CCM.P-K15 and CCM.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-3} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-3} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-2} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-2} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 0.3 Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 9.0E-03 | 3.9E-02 | ||
CENAM | -9.3E-03 | 7.7E-02 | ||
NIST | -2.0E-04 | 2.0E-02 | ||
INRIM | -4.0E-04 | 2.9E-02 | ||
PTB | -6.5E-03 | 2.7E-02 | ||
METAS | 2.0E-03 | 1.7E-02 | ||
NIM | 9.0E-04 | 1.5E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 6.5E-03 | 1.7E-02 | ||
CENAM | 5.4E-03 | 2.5E-02 | ||
NIST | -1.9E-03 | 7.0E-03 | ||
INRIM | 6.0E-04 | 1.9E-02 | ||
PTB | -3.3E-03 | 1.3E-02 | ||
METAS | 1.1E-03 | 9.2E-03 | ||
NIM | 1.2E-03 | 7.1E-03 | ||
NIMT | -9.8E-03 | 3.2E-02 | ||
UME | -6.4E-03 | 1.7E-02 | ||
IMT | -4.4E-03 | 1.6E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-4} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 5.8E-03 | 1.1E-02 | ||
KRISS | -1.9E-03 | 2.0E-02 | ||
CENAM | 2.8E-03 | 1.3E-02 | ||
NIST | -4.0E-04 | 3.5E-03 | ||
INRIM | -7.0E-04 | 1.5E-02 | ||
PTB | -3.2E-03 | 1.0E-02 | ||
METAS | -9.0E-04 | 9.7E-03 | ||
NIM | 1.2E-03 | 4.7E-03 | ||
NIMT | -9.0E-03 | 1.9E-02 | ||
UME | -4.1E-03 | 1.5E-02 | ||
IMT | -3.6E-03 | 1.4E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-3} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 6.5E-03 | 1.1E-02 | ||
KRISS | -1.2E-03 | 2.0E-02 | ||
CENAM | 2.0E-04 | 1.1E-02 | ||
NIST | -4.0E-04 | 3.9E-03 | ||
INRIM | 0.0E+00 | 1.2E-02 | ||
PTB | -2.9E-03 | 9.9E-03 | ||
METAS | -3.0E-04 | 5.7E-03 | ||
NIM | 1.0E-03 | 3.9E-03 | ||
NIMT | -8.6E-03 | 1.8E-02 | ||
UME | -4.8E-03 | 1.4E-02 | ||
IMT | -2.8E-03 | 1.4E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-3} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 6.0E-03 | 1.1E-02 | ||
KRISS | -2.9E-03 | 2.0E-02 | ||
CENAM | 1.0E-04 | 5.4E-03 | ||
NIST | -2.0E-04 | 3.4E-03 | ||
INRIM | -5.0E-04 | 1.0E-02 | ||
PTB | -1.6E-03 | 9.5E-03 | ||
METAS | -5.0E-04 | 5.9E-03 | ||
NIM | 6.0E-04 | 3.6E-03 | ||
NIMT | -7.0E-03 | 1.6E-02 | ||
UME | -4.6E-03 | 1.4E-02 | ||
IMT | -2.5E-03 | 1.3E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-2} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 6.0E-03 | 1.0E-02 | ||
KRISS | -6.2E-03 | 2.1E-02 | ||
CENAM | -4.0E-04 | 4.9E-03 | ||
NIST | -2.0E-04 | 3.5E-03 | ||
INRIM | -4.0E-04 | 1.0E-02 | ||
PTB | -2.0E-03 | 9.8E-03 | ||
METAS | 3.0E-04 | 5.9E-03 | ||
NIM | 1.3E-03 | 3.4E-03 | ||
NIMT | -7.2E-03 | 1.8E-02 | ||
UME | -3.7E-03 | 1.4E-02 | ||
IMT | -1.4E-03 | 1.3E-02 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-2} Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 6.5E-03 | 1.1E-02 | ||
KRISS | -1.3E-02 | 2.4E-02 | ||
CENAM | -1.0E-04 | 5.0E-03 | ||
NIST | 6.0E-04 | 3.8E-03 | ||
INRIM | -3.0E-04 | 7.5E-03 | ||
PTB | -2.6E-03 | 6.1E-03 | ||
METAS | 3.0E-04 | 6.1E-03 | ||
NIM | 1.3E-03 | 2.8E-03 | ||
NIMT | -7.5E-03 | 1.3E-02 | ||
UME | -4.1E-03 | 1.0E-02 | ||
IMT | -2.4E-03 | 9.3E-03 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 0.3 Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 5.3E-03 | 1.1E-02 | ||
CENAM | -1.1E-03 | 5.0E-03 | ||
NIST | 1.0E-03 | 4.1E-03 | ||
INRIM | -8.0E-04 | 6.7E-03 | ||
PTB | -2.7E-03 | 5.5E-03 | ||
METAS | 9.0E-04 | 4.3E-03 | ||
NIM | 3.0E-04 | 3.1E-03 | ||
NIMT | -6.3E-03 | 1.3E-02 | ||
UME | -4.1E-03 | 9.3E-03 | ||
IMT | -1.7E-03 | 8.8E-03 |
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 Pa
Degrees of equivalence represented by D_{i} = (x_{I} - x_{R})/x_{R }and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in non-dimensional units.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
Pa/Pa |
Pa/Pa |
Pa/Pa |
||
NMIJ | 4.6E-03 | 1.1E-02 | ||
CENAM | 2.0E-04 | 4.1E-03 | ||
NIST | 1.3E-03 | 3.1E-03 | ||
INRIM | -3.2E-03 | 7.5E-03 | ||
PTB | -3.6E-03 | 5.9E-03 | ||
METAS | 1.0E-04 | 3.9E-03 | ||
NIM | 4.0E-04 | 3.0E-03 | ||
NIMT | -5.6E-03 | 1.3E-02 | ||
UME | -4.0E-03 | 9.6E-03 | ||
IMT | -1.7E-03 | 9.1E-03 |
Metrology area, Sub-field | Mass and related quantities, Pressure |
Description | Comparison of absolute pressure in the range from 0.0003 Pa to 1 Pa |
Time of measurements | 2017 |
Status | Approved for equivalence |
Final Reports of the comparisons | |
References | |
Measurand | Pressure 0.0003 Pa, 0.0009 Pa, 0.003 Pa, 0.009 Pa, 0.03 Pa, 0.09 Pa, 0.3 Pa and 1 Pa |
Parameters | Nitrogen gas with 99.99% purity used for the calibration Temperature at [23 ± 2] °C |
Transfer device | Two spinning rotor gauges (SRG) |
Comparison type | Key Comparison |
Consultative Committee | CCM (Consultative Committee for Mass and Related Quantities) |
Conducted by | EURAMET (European Association of National Metrology Institutes) |
RMO Internal Identifier | EURAMET.M.1405 |
Comments | Results published on 2 December 2020 Results at 1 Pa are also linked to CCM.P-K4 |
Pilot institute |
PTB
Physikalisch-Technische Bundesanstalt Germany |
Contact person | K. Jousten +49 (0) 30 34 81 7262 |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCM.P-K15 and CCM.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 x 10^{-4} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-4} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-4} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-3} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-3} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-2} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-2} Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 0.3 Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 Pa
CCM.P-K15
The key comparison reference value, x_{R}, is adopted as the nominal value of the pressure. The standard uncertainty of the key comparison reference value, u_{R}, is calculated according to Eq. (40) of the CCM.P-K15 Final Report, at each pressure.
For each pressure, the degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} , its relative expanded uncertainty (k = 2) calculated according to Eq. (41) of the CCM.P-K15 Final Report.
Linking CCM.P-K15.1 to CCM.P-K15
The results of CCM.P-K15.1 were linked to CCM.P-K15 via METAS who participated in both comparisons.
Linking EURAMET.M.P-K15.1 to CCM.P-K15
The results of EURAMET.M.P-K15.1 were linked to CCM.P-K15 via PTB who participated in both comparisons.
CCM.P-K15 and CCM.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 x 10^{-4} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-4} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-4} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-3} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-3} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 3.0 x 10^{-2} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 9.0 x 10^{-2} Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 0.3 Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.
CCM.P-K15, CCM.P-K15.1 and EURAMET.M.P-K15.1
MEASURAND Pressure
NOMINAL VALUE 1.0 Pa
CCM.P-K15
The laboratory measurement obtained in CCM.P-K15 are listed in Sec. 5 of the CCM.P-K15 Final report.
CCM.P-K15.1
The laboratory measurement obtained in CCM.P-K15.1 are listed in Sec. 5 of the CCM.P-K15.1 Final report.
EURAMET.M.P-K15.1
The laboratory measurement obtained in ERURAMET.M.P-K15.1 are listed in Sec. of the EURAMET.M.P-K15.1 Final report.