CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 50 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 100 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 150 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 200 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 250 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 300 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 350 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 400 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 450 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 500 MPa
Graph of equivalence
Red diamonds: participants in CCM.PK13
Blue triangles: participants in APMP.M.PK13
Green squares: participants in EURAMET.M.PK13
Orange circles: participants in EURAMET.M.PK7
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 50 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 100 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 150 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 200 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 250 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13 and EURAMET.M.PK13
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 300 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 350 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 400 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 450 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 500 MPa
Degrees of equivalence relative to the CCM.PK3 key comparison reference value: D_{i} and U_{i} (k = 2), both expressed in 10^{6}
Red: participants in CCM.PK13
Blue: participants in APMP.M.PK13
Green: participants in EURAMET.M.PK13
Orange: participants in EURAMET.M.PK7
Results are presented under A4 printable format in Summary Results (PDF file).
Metrology area, Subfield  Mass and related quantities, Pressure 
Description  Hydraulic gauge pressure 
Time of measurements  2008  2010 
Status  Approved for equivalence 
Final Reports of the comparisons  
Measurand  Effective area of a pistoncylinder unit 
Parameters  Hydraulic gauge pressure: 50 MPa to 500 MPa 
Transfer device  Pistoncylinder assembly in a pressure balance 
Comparison type  Key Comparison 
Consultative Committee  CCM (Consultative Committee for Mass and Related Quantities) 
Conducted by  CCM (Consultative Committee for Mass and Related Quantities) 
Comments 
Results published on 09 May 2012 APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7 results are linked to those of CCM.PK13 Hydraulic gauge pressure

Pilot institute 
PTB
PhysikalischTechnische Bundesanstalt Germany 
Contact person  Wladimir Sabuga +49 (0) 531 592 3230 
Pilot laboratory  

PTB 
PhysikalischTechnische Bundesanstalt, Germany, EURAMET 
CENAM 
Centro Nacional de Metrologia, Mexico, SIM 
LNE 
Laboratoire national de métrologie et d'essais, France, EURAMET 
NIM 
National Institute of Metrology, China, APMP 
NIST 
National Institute of Standards and Technology, United States, SIM 
NMIJ AIST 
National Metrology Institute of Japan, Japan, APMP 
NPLI 
CSIR National Physical Laboratory of India, India, APMP 
This page proposes printout on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 50 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 100 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 150 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 200 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 250 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 300 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 350 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 400 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 450 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 500 MPa
The key comparison reference value, x_{R}, is calculated as the median of the participants' values. The standard uncertainty of the key comparison reference value, u_{R}, is calculated by the formula u_{R} = 1.858 MED x_{i}  x_{R} / (n  1)^{1/2}, where MED means median and n is the number of participants. 
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} = (x_{i}  x_{R}) / x_{R}, the relative deviation from the reference value, and U_{i} = 2(u_{i}^{2} + u_{DC}^{2} + u_{R}^{2})^{1/2} / x_{R}, its expanded uncertainty (k = 2). 
When required, the degree of equivalence between two laboratories i and j is given by a pair of terms: No pairwise degrees of equivalence are displayed here, but they can be found in the relevant Final Report. 
The results of the participants in APMP.M.PK13 are linked to those of CCM.PK13 through the results of the two linking laboratories, NMIJ and NPLI, as explained in Section 7 of the APMP.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK13 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 6 of the EURAMET.M.PK13 Final Report.
The results of the participants in EURAMET.M.PK7 are linked to those of CCM.PK13 through the results of the linking laboratory, PTB, as explained in Section 9 of the EURAMET.M.PK7 Final Report.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 50 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 0.5 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
1.961187 
17 
Jan 2009 
CENAM  1.961171  85  Mar 2009 
NPLI  1.961104  44  May 2009 
NIM  1.961098  17  Jun 2009 
NMIJ  1.961152  17  Oct 2009 
LNE  1.961191  11  Feb 2010 
PTB  1.961148  15  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 100 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 0.5 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
1.961285 
17 
Jan 2009 
CENAM  1.961258  85  Mar 2009 
NPLI  1.961244  44  May 2009 
NIM  1.961175  17  Jun 2009 
NMIJ  1.961269  17  Oct 2009 
LNE  1.961273  13  Feb 2010 
PTB  1.961261  15  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 150 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 0.6 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
1.961409 
17 
Jan 2009 
CENAM  1.961367  85  Mar 2009 
NPLI  1.961376  44  May 2009 
NIM  1.961280  17  Jun 2009 
NMIJ  1.961386  19  Oct 2009 
LNE  1.961373  16  Feb 2010 
PTB  1.961378  16  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 200 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 0.2 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
1.961519 
17 
Jan 2009 
CENAM  1.961481  85  Mar 2009 
NPLI  1.961493  44  May 2009 
NIM  1.961381  18  Jun 2009 
NMIJ  1.961495  20  Oct 2009 
LNE  1.961474  18  Feb 2010 
PTB  1.961486  17  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 250 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 0.4 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
1.961607 
17 
Jan 2009 
CENAM  1.961589  85  Mar 2009 
NPLI  1.961603  44  May 2009 
NIM  1.961477  19  Jun 2009 
NMIJ  1.961597  22  Oct 2009 
LNE  1.961574  20  Feb 2010 
PTB  1.961589  18  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 300 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 1.2 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
 
 
 
CENAM  1.961699  85  Mar 2009 
NPLI  1.961705  44  May 2009 
NIM  1.961564  20  Jun 2009 
NMIJ  1.961692  24  Oct 2009 
LNE  1.961671  23  Feb 2010 
PTB  1.961685  20  Mar 2010 
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 350 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 1.4 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
 
 
 
CENAM  1.961802  85  Mar 2009 
NPLI  1.961802  44  May 2009 
NIM  1.961646  21  Jun 2009 
NMIJ  1.961782  27  Oct 2009 
LNE  1.961766  25  Feb 2010 
PTB  1.961777  22  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 400 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 1.6 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
 
 
 
CENAM  1.961905  85  Mar 2009 
NPLI  1.961887  44  May 2009 
NIM  1.961729  22  Jun 2009 
NMIJ  1.961871  30  Oct 2009 
LNE  1.961856  28  Feb 2010 
PTB  1.961866  24  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 450 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 2.1 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
 
 
 
CENAM  1.962013  85  Mar 2009 
NPLI  1.961987  44  May 2009 
NIM  1.961774  23  Jun 2009 
NMIJ  1.961956  33  Oct 2009 
LNE  1.961945  30  Feb 2010 
PTB  1.961953  26  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.
CCM.PK13, APMP.M.PK13, EURAMET.M.PK13 and EURAMET.M.PK7
MEASURAND : Effective area A' (p', 20 °C) of pistoncylinder unit DHBudenberg 4603 mounted in a DHBudenberg pressure balance model 5306 determined from comparisons against participants primary pressure standards
NOMINAL VALUE : 2 mm^{2}
NOMINAL PRESSURE : p' = 500 MPa
x_{i} : result of measurement of A' (p', 20 °C) taken as the arithmetic average of 10 values obtained by laboratory i and corrected for drift
u_{i} : combined standard uncertainty of x_{i} as reported by laboratory i (without including uncertainty of the drift correction)
u_{DC} : standard uncertainty of the drift correction of x_{i}
u_{DC} / x_{i} = 2.2 x 10^{6}
Lab i  x_{i} / mm^{2} 
u_{i} / x_{i} / 10^{6} 
Date of measurement 
NIST 
 
 
 
CENAM  1.962120  85  Mar 2009 
NPLI  1.962093  44  May 2009 
NIM  1.961853  25  Jun 2009 
NMIJ  1.962041  36  Oct 2009 
LNE  1.962040  33  Feb 2010 
PTB  1.962040  28  Mar 2010 
The results of the participating laboratories in key comparison APMP.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2010 and 2013.
The results of the participating laboratories in key comparison EURAMET.M.PK13 are given in Section 6 of the Final Report.
Measurements were carried out between 2009 and 2011.
The results of the participating laboratories in key comparison EURAMET.M.PK7 are given in Section 7 of the Final Report.
Measurements were carried out between 2005 and 2007.