MEASURAND : Spectral responsivity
WAVELENGTH : 900 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 950 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1000 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1050 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1100 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1150 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1200 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1250 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1300 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1350 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1400 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1450 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1500 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1550 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 1600 nm
Degrees of equivalence relative to the key comparison reference value, D_{i} and U_{i} its expanded uncertainty (k = 2), both expressed in %
MEASURAND : Spectral responsivity
WAVELENGTH : 900 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.3 | 0.6 |
NMISA | 0.1 | 2.6 |
NMIA | -0.8 | 0.4 |
HUT | -0.4 | 3.1 |
IFA-CSIC | 1.9 | 1.3 |
KRISS | 0.1 | 0.8 |
NIM | 3.5 | 1.4 |
NIST | 0.0 | 0.4 |
VSL | 0.7 | 1.2 |
NPL | 0.4 | 0.4 |
NRC | 0.7 | 0.4 |
MKEH | 0.0 | 2.8 |
PTB | -0.1 | 0.4 |
SMU | 0.1 | 0.6 |
VNIIOFI | 0.3 | 1.2 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 950 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.2 | 0.5 |
NMISA | 0.1 | 2.6 |
NMIA | -0.3 | 0.3 |
HUT | 0.9 | 3.6 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.1 | 0.8 |
NIM | 1.3 | 0.9 |
NIST | 0.1 | 1.3 |
VSL | 0.2 | 0.9 |
NPL | 0.5 | 0.3 |
NRC | 0.0 | 0.3 |
MKEH | -0.4 | 2.8 |
PTB | -0.2 | 0.3 |
SMU | 0.5 | 0.5 |
VNIIOFI | -1.0 | 1.0 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1000 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.1 | 0.5 |
NMISA | 0.1 | 2.6 |
NMIA | -0.4 | 0.3 |
HUT | -0.1 | 3.5 |
IFA-CSIC | 1.3 | 1.2 |
KRISS | 0.3 | 0.8 |
NIM | 1.3 | 0.9 |
NIST | -0.2 | 0.9 |
VSL | 0.5 | 0.9 |
NPL | 0.1 | 0.3 |
NRC | 0.1 | 0.2 |
MKEH | -0.2 | 2.8 |
PTB | 0.1 | 0.3 |
SMU | 0.4 | 0.4 |
VNIIOFI | -0.8 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1050 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.1 | 0.5 |
NMISA | 0.2 | 2.2 |
NMIA | -0.3 | 0.3 |
HUT | 0.5 | 3.5 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.0 | 1.0 |
NIM | 0.8 | 0.8 |
NIST | 0.2 | 0.9 |
VSL | 0.3 | 0.8 |
NPL | 0.1 | 0.3 |
NRC | 0.1 | 0.2 |
MKEH | 0.6 | 3.5 |
PTB | 0.1 | 0.3 |
SMU | 0.2 | 0.4 |
VNIIOFI | -0.9 | 1.3 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1100 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.0 | 0.5 |
NMISA | -0.1 | 2.2 |
NMIA | -0.3 | 0.3 |
HUT | -0.1 | 3.5 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.1 | 1.0 |
NIM | 0.3 | 0.9 |
NIST | 0.0 | 0.5 |
VSL | 0.3 | 0.6 |
NPL | 0.2 | 0.3 |
NRC | 0.0 | 0.2 |
MKEH | 0.8 | 3.3 |
PTB | 0.1 | 0.3 |
SMU | 0.2 | 0.4 |
VNIIOFI | -1.2 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1150 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -1.0 | 0.5 |
NMISA | 0.3 | 2.2 |
NMIA | -0.1 | 0.3 |
HUT | 0.2 | 3.5 |
IFA-CSIC | 1.4 | 1.2 |
KRISS | 0.1 | 1.0 |
NIM | 0.0 | 0.8 |
NIST | 0.1 | 0.8 |
VSL | 0.2 | 0.5 |
NPL | 0.2 | 0.3 |
NRC | 0.0 | 0.2 |
MKEH | 0.7 | 3.5 |
PTB | 0.2 | 0.2 |
SMU | 0.2 | 0.4 |
VNIIOFI | -1.5 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1200 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.7 | 0.5 |
NMISA | 0.0 | 2.2 |
NMIA | -0.3 | 0.2 |
HUT | 0.4 | 3.5 |
IFA-CSIC | 1.6 | 1.2 |
KRISS | 0.1 | 1.0 |
NIM | -0.3 | 0.9 |
NIST | 0.0 | 1.5 |
VSL | 0.1 | 0.5 |
NPL | 0.3 | 0.3 |
NRC | -0.1 | 0.2 |
MKEH | 0.7 | 3.4 |
PTB | 0.2 | 0.2 |
SMU | 0.2 | 0.4 |
VNIIOFI | -1.9 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1250 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.8 | 0.5 |
NMISA | 0.1 | 2.0 |
NMIA | -0.4 | 0.2 |
HUT | 0.0 | 3.5 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.0 | 1.0 |
NIM | -0.4 | 0.8 |
NIST | 0.1 | 0.9 |
VSL | 0.2 | 0.4 |
NPL | 0.2 | 0.3 |
NRC | 0.0 | 0.2 |
MKEH | 0.3 | 3.4 |
PTB | 0.3 | 0.2 |
SMU | 0.2 | 0.4 |
VNIIOFI | -2.0 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1300 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.8 | 0.5 |
NMISA | 0.2 | 2.0 |
NMIA | -0.3 | 0.2 |
HUT | 0.0 | 3.5 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.2 | 1.0 |
NIM | -0.3 | 0.8 |
NIST | 0.1 | 0.9 |
VSL | 0.2 | 0.4 |
NPL | 0.2 | 0.3 |
NRC | 0.0 | 0.1 |
MKEH | 0.5 | 3.4 |
PTB | 0.1 | 0.2 |
SMU | 0.2 | 0.4 |
VNIIOFI | -2.2 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1350 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.6 | 0.5 |
NMISA | 0.8 | 2.0 |
NMIA | -0.2 | 0.2 |
HUT | 0.5 | 3.5 |
IFA-CSIC | 1.4 | 1.2 |
KRISS | -0.1 | 1.0 |
NIM | -0.5 | 0.8 |
NIST | 0.3 | 0.9 |
VSL | 0.1 | 0.4 |
NPL | 0.3 | 0.3 |
NRC | 0.0 | 0.2 |
MKEH | 0.5 | 3.4 |
PTB | 0.2 | 0.2 |
SMU | 0.1 | 0.4 |
VNIIOFI | -2.6 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1400 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.6 | 0.5 |
NMISA | 1.0 | 2.0 |
NMIA | -0.3 | 0.3 |
HUT | -0.2 | 3.5 |
IFA-CSIC | 1.4 | 1.1 |
KRISS | -0.1 | 1.0 |
NIM | -0.5 | 0.8 |
NIST | 0.5 | 1.2 |
VSL | 0.2 | 0.4 |
NPL | 0.2 | 0.3 |
NRC | 0.0 | 0.2 |
MKEH | 0.8 | 3.3 |
PTB | 0.2 | 0.2 |
SMU | 0.1 | 0.4 |
VNIIOFI | -1.2 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1450 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.8 | 0.5 |
NMISA | 1.4 | 2.0 |
NMIA | -0.3 | 0.2 |
HUT | -0.1 | 3.5 |
IFA-CSIC | 1.5 | 1.2 |
KRISS | 0.1 | 1.0 |
NIM | -0.7 | 0.9 |
NIST | 0.6 | 0.9 |
VSL | 0.2 | 0.4 |
NPL | 0.2 | 0.4 |
NRC | 0.2 | 0.2 |
MKEH | 0.9 | 3.4 |
PTB | 0.0 | 0.2 |
SMU | 0.2 | 0.4 |
VNIIOFI | -2.8 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1500 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.7 | 0.5 |
NMISA | 1.7 | 2.0 |
NMIA | -0.3 | 0.3 |
HUT | 0.1 | 3.5 |
IFA-CSIC | 1.3 | 1.2 |
KRISS | 0.3 | 1.0 |
NIM | -0.7 | 1.0 |
NIST | 0.9 | 1.0 |
VSL | 0.3 | 0.4 |
NPL | 0.3 | 0.3 |
NRC | 0.1 | 0.2 |
MKEH | 0.6 | 3.4 |
PTB | 0.1 | 0.2 |
SMU | 0.1 | 0.4 |
VNIIOFI | -3.2 | 1.2 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1550 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.7 | 0.5 |
NMISA | 2.4 | 2.0 |
NMIA | -0.4 | 0.3 |
HUT | -0.1 | 3.5 |
IFA-CSIC | 1.4 | 1.2 |
KRISS | 0.0 | 1.0 |
NIM | -0.9 | 1.1 |
NIST | 0.5 | 1.1 |
VSL | 0.3 | 0.8 |
NPL | 0.2 | 0.3 |
NRC | 0.1 | 0.2 |
MKEH | -2.7 | 5.3 |
PTB | 0.3 | 0.2 |
SMU | 0.3 | 0.4 |
VNIIOFI | -3.6 | 1.2 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
MEASURAND : Spectral responsivity
WAVELENGTH : 1600 nm
Lab i | D_{i} | U_{i} |
/ % | / % | |
LNE | -0.6 | 0.5 |
NMISA | 3.2 | 2.0 |
NMIA | -0.3 | 0.3 |
HUT | -0.6 | 3.5 |
IFA-CSIC | 1.6 | 1.2 |
KRISS | 0.3 | 1.1 |
NIM | -0.9 | 1.2 |
NIST | 0.7 | 1.3 |
VSL | 0.4 | 1.4 |
NPL | 0.2 | 0.4 |
NRC | 0.4 | 0.2 |
MKEH | -2.0 | 6.0 |
PTB | 0.4 | 0.3 |
SMU | 0.2 | 0.4 |
VNIIOFI | -4.0 | 1.1 |
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
Metrology area, Sub-field | Photometry and Radiometry, Radiometry |
Description | Spectral responsivity |
Time of measurements | 2010 - 2012 |
Status | Approved for equivalence |
References | |
Measurand | Spectral responsivity |
Parameters | Wavelength from 900 nm to 1600 nm |
Transfer device | Germanium photodiodes |
Comparison type | Key Comparison |
Consultative Committee | CCPR (Consultative Committee for Photometry and Radiometry) |
Conducted by | EURAMET (European Association of National Metrology Institutes) |
Comments | CCPR-K2.a-2003 in the report refers to CCPR-K2.a. |
Pilot institute |
VSL
VSL Netherlands |
Contact person | Steven van den Berg svdberg@vsl.nl +31 15 269 1586 |
Pilot laboratory | |
---|---|
VSL |
VSL, Netherlands, EURAMET |
CMI |
Czech Metrology Institute, Czechia, EURAMET |
GUM |
Glowny Urzad Miar, Central Office of Measures, Poland, EURAMET |
JV |
Justervesenet, Norway, EURAMET |
NPL |
National Physical Laboratory, United Kingdom, EURAMET |
SP |
Technical Research Institute of Sweden from 2017 Research Institutes of Sweden AB, Sweden, EURAMET |
UME |
TÜBITAK Ulusal Metroloji Enstitüsü, Turkey, EURAMET |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
MEASURAND : Spectral responsivity
WAVELENGTH : 900 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 950 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1000 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1050 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1100 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1150 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1200 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1250 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1300 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1350 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1400 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1450 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1500 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1550 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 1600 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). |
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: D_{ij} and U_{ij}, its expanded uncertainty (k = 2), both expressed in %. |
MEASURAND : Spectral responsivity
WAVELENGTH : 900 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 950 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1000 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1050 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1100 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1150 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1200 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1250 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1300 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1350 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1400 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1450 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1500 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1550 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.
MEASURAND : Spectral responsivity
WAVELENGTH : 1600 nm
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
Examination and correction of the data are explained in Section 5.
The data analysis in given in Section 6 and is based on the calculation of the mean relative differences between each participant and the Pilot Laboratory, NIST, weighted by the transfer uncertainty of the measurements of each photodiode.