CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 900 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 950 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1000 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1050 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1100 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1150 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1200 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1250 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1300 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1350 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1400 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1450 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1500 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1550 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1600 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 900 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.34 | 0.56 | ||
NMISA | 0.14 | 2.61 | ||
NMIA | -0.85 | 0.40 | ||
HUT | -0.44 | 3.13 | ||
IFA-CSIC | 1.93 | 1.28 | ||
KRISS | 0.14 | 0.83 | ||
NIM | 3.51 | 1.42 | ||
NIST | -0.03 | 0.39 | ||
VSL | 0.66 | 1.17 | ||
NPL | 0.37 | 0.43 | ||
NRC | 0.66 | 0.41 | ||
MKEH | 0.00 | 2.80 | ||
PTB | -0.12 | 0.41 | ||
SMU | 0.11 | 0.62 | ||
VNIIOFI | 0.33 | 1.22 | ||
CMI | -0.38 | 1.08 | ||
GUM | -2.27 | 1.03 | ||
JV | -0.14 | 1.22 | ||
RISE | -0.08 | 1.76 | ||
UME | 0.20 | 1.89 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 950 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.2 | 0.5 | ||
NMISA | 0.1 | 2.6 | ||
NMIA | -0.3 | 0.3 | ||
HUT | 0.9 | 3.6 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.1 | 0.8 | ||
NIM | 1.3 | 0.9 | ||
NIST | 0.1 | 1.3 | ||
VSL | 0.2 | 0.9 | ||
NPL | 0.5 | 0.3 | ||
NRC | 0.0 | 0.3 | ||
MKEH | -0.4 | 2.8 | ||
PTB | -0.2 | 0.3 | ||
SMU | 0.5 | 0.5 | ||
VNIIOFI | -1.0 | 1.0 | ||
CMI | -0.38 | 1.02 | ||
GUM | -0.19 | 0.94 | ||
JV | -0.39 | 0.85 | ||
RISE | -0.70 | 1.70 | ||
UME | 0.27 | 1.88 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1000 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.1 | 0.5 | ||
NMISA | 0.1 | 2.6 | ||
NMIA | -0.4 | 0.3 | ||
HUT | -0.1 | 3.5 | ||
IFA-CSIC | 1.3 | 1.2 | ||
KRISS | 0.3 | 0.8 | ||
NIM | 1.3 | 0.9 | ||
NIST | -0.2 | 0.9 | ||
VSL | 0.5 | 0.9 | ||
NPL | 0.1 | 0.3 | ||
NRC | 0.1 | 0.2 | ||
MKEH | -0.2 | 2.8 | ||
PTB | 0.1 | 0.3 | ||
SMU | 0.4 | 0.4 | ||
VNIIOFI | -0.8 | 1.1 | ||
CMI | 0.1 | 0.5 | ||
GUM | 0.5 | 0.7 | ||
JV | 0.0 | 0.9 | ||
RISE | 0.1 | 1.7 | ||
UME | 1.6 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1050 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.1 | 0.5 | ||
NMISA | 0.2 | 2.2 | ||
NMIA | -0.3 | 0.3 | ||
HUT | 0.5 | 3.5 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.0 | 1.0 | ||
NIM | 0.8 | 0.8 | ||
NIST | 0.2 | 0.9 | ||
VSL | 0.3 | 0.8 | ||
NPL | 0.1 | 0.3 | ||
NRC | 0.1 | 0.2 | ||
MKEH | 0.6 | 3.5 | ||
PTB | 0.1 | 0.3 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -0.9 | 1.3 | ||
CMI | 0.2 | 0.5 | ||
GUM | 0.5 | 0.8 | ||
JV | 0.2 | 0.9 | ||
RISE | 0.6 | 1.7 | ||
UME | 1.4 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1100 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.0 | 0.5 | ||
NMISA | -0.1 | 2.2 | ||
NMIA | -0.3 | 0.3 | ||
HUT | -0.1 | 3.5 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.1 | 1.0 | ||
NIM | 0.3 | 0.9 | ||
NIST | 0.0 | 0.5 | ||
VSL | 0.3 | 0.6 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.0 | 0.2 | ||
MKEH | 0.8 | 3.3 | ||
PTB | 0.1 | 0.3 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -1.2 | 1.1 | ||
CMI | 0.3 | 0.5 | ||
GUM | 0.9 | 0.7 | ||
JV | 0.2 | 0.9 | ||
RISE | 1.2 | 1.7 | ||
UME | 1.7 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1150 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -1.0 | 0.5 | ||
NMISA | 0.3 | 2.2 | ||
NMIA | -0.1 | 0.3 | ||
HUT | 0.2 | 3.5 | ||
IFA-CSIC | 1.4 | 1.2 | ||
KRISS | 0.1 | 1.0 | ||
NIM | 0.0 | 0.8 | ||
NIST | 0.1 | 0.8 | ||
VSL | 0.2 | 0.5 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.0 | 0.2 | ||
MKEH | 0.7 | 3.5 | ||
PTB | 0.2 | 0.2 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -1.5 | 1.1 | ||
CMI | 0.4 | 0.5 | ||
GUM | 1.0 | 0.8 | ||
JV | 0.4 | 0.8 | ||
RISE | 0.9 | 1.7 | ||
UME | 2.3 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1200 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.7 | 0.5 | ||
NMISA | 0.0 | 2.2 | ||
NMIA | -0.3 | 0.2 | ||
HUT | 0.4 | 3.5 | ||
IFA-CSIC | 1.6 | 1.2 | ||
KRISS | 0.1 | 1.0 | ||
NIM | -0.3 | 0.9 | ||
NIST | 0.0 | 1.5 | ||
VSL | 0.1 | 0.5 | ||
NPL | 0.3 | 0.3 | ||
NRC | -0.1 | 0.2 | ||
MKEH | 0.7 | 3.4 | ||
PTB | 0.2 | 0.2 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -1.9 | 1.1 | ||
CMI | 0.1 | 0.5 | ||
GUM | 0.6 | 0.8 | ||
JV | 0.2 | 0.9 | ||
RISE | 0.5 | 1.7 | ||
UME | 2.5 | 1.8 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1250 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.8 | 0.5 | ||
NMISA | 0.1 | 2.0 | ||
NMIA | -0.4 | 0.2 | ||
HUT | 0.0 | 3.5 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.0 | 1.0 | ||
NIM | -0.4 | 0.8 | ||
NIST | 0.1 | 0.9 | ||
VSL | 0.2 | 0.4 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.0 | 0.2 | ||
MKEH | 0.3 | 3.4 | ||
PTB | 0.3 | 0.2 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -2.0 | 1.1 | ||
CMI | 0.2 | 0.5 | ||
GUM | 0.7 | 0.7 | ||
JV | 0.3 | 0.8 | ||
RISE | 0.3 | 1.7 | ||
UME | 2.9 | 1.8 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1300 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.8 | 0.5 | ||
NMISA | 0.2 | 2.0 | ||
NMIA | -0.3 | 0.2 | ||
HUT | 0.0 | 3.5 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.2 | 1.0 | ||
NIM | -0.3 | 0.8 | ||
NIST | 0.1 | 0.9 | ||
VSL | 0.2 | 0.4 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.0 | 0.1 | ||
MKEH | 0.5 | 3.4 | ||
PTB | 0.1 | 0.2 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -2.2 | 1.1 | ||
CMI | 0.2 | 0.5 | ||
GUM | 0.6 | 0.7 | ||
JV | 0.3 | 0.8 | ||
RISE | 0.3 | 1.7 | ||
UME | 2.8 | 1.8 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1350 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.6 | 0.5 | ||
NMISA | 0.8 | 2.0 | ||
NMIA | -0.2 | 0.2 | ||
HUT | 0.5 | 3.5 | ||
IFA-CSIC | 1.4 | 1.2 | ||
KRISS | -0.1 | 1.0 | ||
NIM | -0.5 | 0.8 | ||
NIST | 0.3 | 0.9 | ||
VSL | 0.1 | 0.4 | ||
NPL | 0.3 | 0.3 | ||
NRC | 0.0 | 0.2 | ||
MKEH | 0.5 | 3.4 | ||
PTB | 0.2 | 0.2 | ||
SMU | 0.1 | 0.4 | ||
VNIIOFI | -2.6 | 1.1 | ||
CMI | 0.3 | 0.5 | ||
GUM | 0.6 | 0.7 | ||
JV | 0.3 | 0.8 | ||
RISE | 0.3 | 1.7 | ||
UME | 3.0 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1400 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.6 | 0.5 | ||
NMISA | 1.0 | 2.0 | ||
NMIA | -0.3 | 0.3 | ||
HUT | -0.2 | 3.5 | ||
IFA-CSIC | 1.4 | 1.1 | ||
KRISS | -0.1 | 1.0 | ||
NIM | -0.5 | 0.8 | ||
NIST | 0.5 | 1.2 | ||
VSL | 0.2 | 0.4 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.0 | 0.2 | ||
MKEH | 0.8 | 3.3 | ||
PTB | 0.2 | 0.2 | ||
SMU | 0.1 | 0.4 | ||
VNIIOFI | -1.2 | 1.1 | ||
CMI | 0.5 | 0.5 | ||
GUM | 0.9 | 0.8 | ||
JV | 0.3 | 0.9 | ||
RISE | 0.9 | 1.7 | ||
UME | 2.7 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1450 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.8 | 0.5 | ||
NMISA | 1.4 | 2.0 | ||
NMIA | -0.3 | 0.2 | ||
HUT | -0.1 | 3.5 | ||
IFA-CSIC | 1.5 | 1.2 | ||
KRISS | 0.1 | 1.0 | ||
NIM | -0.7 | 0.9 | ||
NIST | 0.6 | 0.9 | ||
VSL | 0.2 | 0.4 | ||
NPL | 0.2 | 0.4 | ||
NRC | 0.2 | 0.2 | ||
MKEH | 0.9 | 3.4 | ||
PTB | 0.0 | 0.2 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -2.8 | 1.1 | ||
CMI | 0.7 | 0.5 | ||
GUM | 1.2 | 0.7 | ||
JV | 0.5 | 0.8 | ||
RISE | 1.2 | 1.7 | ||
UME | 3.2 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1500 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.7 | 0.5 | ||
NMISA | 1.7 | 2.0 | ||
NMIA | -0.3 | 0.3 | ||
HUT | 0.1 | 3.5 | ||
IFA-CSIC | 1.3 | 1.2 | ||
KRISS | 0.3 | 1.0 | ||
NIM | -0.7 | 1.0 | ||
NIST | 0.9 | 1.0 | ||
VSL | 0.3 | 0.4 | ||
NPL | 0.3 | 0.3 | ||
NRC | 0.1 | 0.2 | ||
MKEH | 0.6 | 3.4 | ||
PTB | 0.1 | 0.2 | ||
SMU | 0.1 | 0.4 | ||
VNIIOFI | -3.2 | 1.2 | ||
CMI | 0.8 | 0.5 | ||
GUM | 1.3 | 0.7 | ||
JV | 0.6 | 0.8 | ||
RISE | 1.5 | 2.1 | ||
UME | 2.7 | 1.9 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1550 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.7 | 0.5 | ||
NMISA | 2.4 | 2.0 | ||
NMIA | -0.4 | 0.3 | ||
HUT | -0.1 | 3.5 | ||
IFA-CSIC | 1.4 | 1.2 | ||
KRISS | 0.0 | 1.0 | ||
NIM | -0.9 | 1.1 | ||
NIST | 0.5 | 1.1 | ||
VSL | 0.3 | 0.8 | ||
NPL | 0.2 | 0.3 | ||
NRC | 0.1 | 0.2 | ||
MKEH | -2.7 | 5.3 | ||
PTB | 0.3 | 0.2 | ||
SMU | 0.3 | 0.4 | ||
VNIIOFI | -3.6 | 1.2 | ||
CMI | 0.6 | 0.6 | ||
GUM | 1.2 | 0.9 | ||
JV | 0.6 | 0.9 | ||
RISE | -1.5 | 2.3 | ||
UME | 1.0 | 2.0 |
RISE, formerly SP
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1600 nm
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
% |
% |
% |
||
LNE | -0.6 | 0.5 | ||
NMISA | 3.2 | 2.0 | ||
NMIA | -0.3 | 0.3 | ||
HUT | -0.6 | 3.5 | ||
IFA-CSIC | 1.6 | 1.2 | ||
KRISS | 0.3 | 1.1 | ||
NIM | -0.9 | 1.2 | ||
NIST | 0.7 | 1.3 | ||
VSL | 0.4 | 1.4 | ||
NPL | 0.2 | 0.4 | ||
NRC | 0.4 | 0.2 | ||
MKEH | -2.0 | 6.0 | ||
PTB | 0.4 | 0.3 | ||
SMU | 0.2 | 0.4 | ||
VNIIOFI | -4.0 | 1.1 | ||
CMI | 0.1 | 0.7 | ||
GUM | 1.7 | 1.2 | ||
JV | 0.1 | 1.1 | ||
RISE | -0.3 | 3.7 | ||
UME | -1.5 | 2.1 |
RISE, formerly SP
Metrology area, Sub-field | Photometry and Radiometry, Radiometry |
Description | Spectral responsivity |
Time of measurements | 2010 - 2012 |
Status | Approved for equivalence |
Final Reports of the comparisons | |
References | |
Measurand | Spectral responsivity |
Parameters | Wavelength from 900 nm to 1600 nm |
Transfer device | Germanium photodiodes |
Comparison type | Key Comparison |
Consultative Committee | CCPR (Consultative Committee for Photometry and Radiometry) |
Conducted by | EURAMET (European Association of National Metrology Institutes) |
RMO Internal Identifier | EURAMET 1116 |
Comments | CCPR-K2.a-2003 in the report refers to CCPR-K2.a. EURAMET.PR-K2.a.2011 was previuously registered as EURAMET.PR-K2.a |
Pilot institute |
VSL
VSL Netherlands |
Contact person | Steven van den Berg svdberg@vsl.nl +31 15 269 1586 |
Pilot laboratory | |
---|---|
VSL |
VSL, Netherlands, EURAMET |
CMI |
Czech Metrology Institute, Czechia, EURAMET |
GUM |
Glowny Urzad Miar, Central Office of Measures, Poland, EURAMET |
JV |
Justervesenet, Norway, EURAMET |
NPL |
National Physical Laboratory, United Kingdom, EURAMET |
SP |
Technical Research Institute of Sweden from 2017 Research Institutes of Sweden AB, Sweden, EURAMET |
UME |
TÜBITAK Ulusal Metroloji Enstitüsü, Turkey, EURAMET |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 900 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 950 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1000 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1050 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1100 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1150 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1200 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1250 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1300 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1350 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1400 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1450 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1500 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1550 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1600 nm
The key comparison reference value is calculated as a weighted mean with cut-off, excluding NMISA and KRISS results (see on page 80 of Section 7 of the CCPR-K2.a Final Report). The value of the cut-off varies with the wavelength and is listed in table 7.17 on page 83 of the Final Report
The degree of equivalence of each laboratory i with respect to the key comparison reference value is given by a pair of terms: D_{i} and U_{i }at a 95 % level of confidenceits expanded uncertainty (k = 2), both expressed in %.
Linking EURAMET.PR-K2.a.2011 to CCPR-K2.a
The results issued from EURAMET.PR-K2.a.2011 were linked to CCPR-K2.a via NPL and VSL who participated in both comparisons.
The linking process is described in Section 5 of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 900 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 950 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1000 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1050 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1100 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1150 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1200 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1250 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1300 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1350 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1400 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1450 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1500 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1550 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.
CCPR-K2.a and EURAMET.PR-K2.a.2011
MEASURAND Spectral responsivity
Wavelength 1600 nm
CCPR-K2.a
The individual measurements of the participating laboratories are given in Section 4 of the CCPR-K2.a Final Report.
EURAMET.PR-K2.a.2011
The individual measurements of the participating laboratories are given in Section 5 and Appendix of the EURAMET.PR-K2.a.2011 Final Report.