CCM.H-K1.c, COOMET.M.H-K1.c, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 240 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 540 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 840 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
CCM.H-K1.c, COOMET.M.H-K1.c, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 240 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
- |
- |
- |
||
INRIM | -1.3 | 1.5 | ||
NIM | -2.8 | 2.2 | ||
KRISS | 2.8 | 6.1 | ||
CMI | -0.4 | 3.1 | ||
GUM | 3.4 | 9.2 | ||
NPL | -0.6 | 3.7 | ||
PTB | -1.2 | 2.7 | ||
KRISS | 0.8 | 2.2 | ||
NIM | -0.9 | 2.2 | ||
SCL | 0.5 | 6.3 | ||
CMS | 0.8 | 3.3 | ||
NIMT | 1.6 | 4.6 | ||
NMIJ | 0.4 | 4.6 | ||
VNIIFTRI | -1.5 | 4.4 | ||
UME | -1.0 | 3.8 |
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 540 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
- |
- |
- |
||
INRIM | -0.06 | 9.09 | ||
INMETRO | 9.43 | 18.48 | ||
NIM | -6.63 | 9.51 | ||
KRISS | 2.90 | 10.98 | ||
CMI | -4.78 | 2.92 | ||
GUM | 2.39 | 21.60 | ||
NPL | -0.65 | 2.68 | ||
PTB | -2.62 | 8.09 | ||
KRISS | -3.3 | 8.3 | ||
NIM | 3.9 | 8.7 | ||
SCL | 6.8 | 17.2 | ||
CMS | -1.0 | 10.4 | ||
NIMT | 1.7 | 13.8 | ||
NMIJ | 2.1 | 13.8 | ||
VNIIFTRI | -2.40 | 12.70 | ||
VNIIFTRI | -3.6 | 10.8 | ||
KazInMetr | 1.0 | 10.0 | ||
BelGIM | -6.9 | 10.4 | ||
NCS IM | -1.9 | 9.6 | ||
UME | -1.5 | 13.6 |
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 840 HV
Degrees of equivalence represented by D_{i} and its expanded uncertainty U_{i} at a 95 % level of confidence.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
- |
- |
- |
||
INRIM | 1.50 | 14.50 | ||
NMIJ | -7.76 | 4.69 | ||
NIM | -3.44 | 7.46 | ||
KRISS | 12.51 | 6.21 | ||
GUM | -1.32 | 25.50 | ||
PTB | -1.49 | 13.93 | ||
KRISS | -9.0 | 16.3 | ||
NIM | 8.3 | 16.9 | ||
SCL | 24.8 | 32.2 | ||
CMS | 2.6 | 18.0 | ||
NIMT | 8.7 | 22.7 | ||
NMIJ | -2.0 | 22.7 | ||
VNIIFTRI | 3.08 | 25.90 | ||
VNIIFTRI | -0.5 | 23.8 | ||
KazInMetr | 11.9 | 22.6 | ||
BelGIM | -3.9 | 22.8 | ||
NSC IM | 7.0 | 22.0 | ||
UME | -1.5 | 24.2 |
Metrology area, Sub-field | Mass and related quantities, Hardness |
Description | Hardness (Vickers 30) |
Time of measurements | 2003 - 2005 |
Status | Approved for equivalence |
Final Reports of the comparisons | |
Measurand | Hardness (Vickers 30): 200 HV, 600 HV and 900 HV |
Transfer device | Three hardness test blocks |
Comparison type | Key Comparison |
Consultative Committee | CCM (Consultative Committee for Mass and Related Quantities) |
Conducted by | APMP (Asia Pacific Metrology Programme) |
RMO Internal Identifier | P-APMP-M.H-K1 |
Comments | Results published on 9 July 2020 APMP.M.H-K1.c results are linked to those of CCM.H-K1.c Hardness Vickers scale HV30 Additional measurements are carried out with a testing force of 98.07 N (Vickers 10) |
Pilot institute |
NMIJ AIST
National Metrology Institute of Japan Japan |
Contact person | S. Takagi +81 298 61 4048 |
Additional |
Pilot laboratory | |
---|---|
NMIJ AIST |
National Metrology Institute of Japan, Japan, APMP |
CMS |
ITRI Center for Measurement Standards, Chinese Taipei, APMP |
KRISS |
Korea Research Institute of Standards and Science, Korea, Republic of, APMP |
NIM |
National Institute of Metrology, China, APMP |
NIMT |
National Institute of Metrology (Thailand), Thailand, APMP |
SCL |
Standards and Calibration Laboratory, Hong Kong, China, APMP |
VMI-STAMEQ |
Viet Nam Metrology Institute, Directorate for Standards and Quality, Viet Nam, APMP |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCM.H-K1.c, COOMET.M.H-K1.c, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 240 HV
The key comparison reference value, x_{R}, and its standard uncertainty, u_{R}, is evaluated by the unified procedure explained in Section 8.2 on page 20 of the CCM.H-K1.c Final Report, based on the arithmetic mean of the participants'' results with exclusion of outliers.
x_{R }=_{ }239.15 and U_{R} = 1.97
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) and U_{i} = 2[abs(u_{i}^{2} - u_{R}^{2})]^{1/2} (where "abs" stands for "absolute value"), its expanded uncertainty (k = 2).
Linking key comparison COOMET.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of PTB in both key comparisons and is explained in Appendix A3 of the COOMET.M.H-K1.c Final Report. It leads to the computation of the degrees of equivalence of VNIIFTRI relative to the CCM.H-K1.c reference values.
Linking key comparison APMP.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of KRISS, NIM, and NMIJ in both key comparisons and is explained in Section 8.1 of the APMP.M.H-K1.c Final Report.
Linking key comparison EURAMET.c to CCM.H-K1.c
The linkage is obtained through the common participation of INRIM in both key comparisons and is explained in Section 11 of the EURAMET.M.H-K1.c Final Report.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 540 HV
The key comparison reference value, x_{R}, and its standard uncertainty, u_{R}, is evaluated by the unified procedure explained in Section 8.2 on page 20 of the CCM.H-K1.c Final Report, based on the arithmetic mean of the participants'' results with exclusion of outliers.
x_{R }=_{ }525.51 and U_{R} = 5.03
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) and U_{i} = 2[abs(u_{i}^{2} - u_{R}^{2})]^{1/2} (where "abs" stands for "absolute value"), its expanded uncertainty (k = 2).
Linking key comparison COOMET.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of PTB in both key comparisons and is explained in Appendix A3 of the COOMET.M.H-K1.c Final Report. It leads to the computation of the degrees of equivalence of VNIIFTRI relative to the CCM.H-K1.c reference values.
Linking key comparison COOMET.M.H-K1 to CCM.H-K1.c
The linkage is obtained through the common participation of PTB in both key comparisons and is explained in Appendix C of the COOMET.M.H-K1 Final Report. It leads to the computation of the degrees of equivalence of participants in COOMET.M.H-K1 relative to the CCM.H-K1.c reference values for 540 HV and 840 HV.
Linking key comparison APMP.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of KRISS, NIM, and NMIJ in both key comparisons and is explained in Section 8.1 of the APMP.M.H-K1.c Final Report.
Linking key comparison EURAMET.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of INRIM in both key comparisons and is explained in Section 111 of the EURAMET.M.H-K1.c Final Report.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 840 HV
The key comparison reference value, x_{R}, and its standard uncertainty, u_{R}, is evaluated by the unified procedure explained in Section 8.2 on page 20 of the CCM.H-K1.c Final Report, based on the arithmetic mean of the participants'' results with exclusion of outliers.
x_{R }=_{ }819.35 and U_{R} = 11.25
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms:
D_{i} = (x_{i} - x_{R}) and U_{i} = 2[abs(u_{i}^{2} - u_{R}^{2})]^{1/2} (where "abs" stands for "absolute value"), its expanded uncertainty (k = 2).
Linking key comparison COOMET.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of PTB in both key comparisons and is explained in Appendix A3 of the COOMET.M.H-K1.c Final Report. It leads to the computation of the degrees of equivalence of VNIIFTRI relative to the CCM.H-K1.c reference values.
Linking key comparison COOMET.M.H-K1 to CCM.H-K1.c
The linkage is obtained through the common participation of PTB in both key comparisons and is explained in Appendix C of the COOMET.M.H-K1 Final Report. It leads to the computation of the degrees of equivalence of participants in COOMET.M.H-K1 relative to the CCM.H-K1.c reference values for 540 HV and 840 HV.
Linking key comparison APMP.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of KRISS, NIM, and NMIJ in both key comparisons and is explained in Section 8.1 of the APMP.M.H-K1.c Final Report.
Linking key comparison EURAMET.M.H-K1.c to CCM.H-K1.c
The linkage is obtained through the common participation of INRIM in both key comparisons and is explained in Section 8.1 of the EURAMET.M.H-K1.c Final Report.
CCM.H-K1.c, COOMET.M.H-K1.c, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 240 HV
CCM.H-K1.c
x_{i} result obtained by laboratory i expressed as the mean value of 8 repetiton measurements
u_{i} combined standard uncertainty of x_{i}
Lab_{i} | x_{i} | u_{i} | Reception | Results available | |
of package | at Pilot Laboratory | ||||
INRIM | 237.83 | 2.1 | Mar 2001 | May 2001 | |
INMETRO | 244.83 | * | May 2001 | Aug 2001 | |
NIST | - | - | Aug 2001 | Sep 2001 | |
NMIJ | 236.11 | * | Sep 2001 | Feb 2003 | |
NIM | 236.40 | 1.7 | Feb 2002 | Mar 2002 | |
KRISS | 241.96 | 3.6 | Mar 2002 | Nov 2002 | |
CMI | 238.73 | 1.2 | May 2002 | Sep 2002 | |
GUM | 242.58 | 5.0 | Sep 2002 | Oct 2002 | |
NPL | 238.55 | 0.7 | Oct 2002 | Feb 2003 | |
PTB | 238.00 | 1.4 | Feb 2003 | Mar 2003 | |
'- : no measurements available | * : no official uncertainty value |
COOMET.M.H-K1.c
The measurement results are listed in Table 1 of the COOMET.M.H-K1.c Final Report.
APMP.M.H-K1.c
The measurement results are listed in Table 7.1 of the APMP.M.H-K1.c Final Report.
EURAMET.M.H-K1.c
The measurement results are listed in Table 5 of the EURAMET.M.H-K1.c Final Report.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 540 HV
CCM.H-K1.c
x_{i} result obtained by laboratory i expressed as the mean value of 8 repetiton measurements
u_{i} combined standard uncertainty of x_{i}
Lab_{i} | x_{i} | u_{i} | Reception | Results available | |
of package | at Pilot Laboratory | ||||
INRIM | 525.45 | 5.0 | Mar 2001 | May 2001 | |
INMETRO | 534.94 | 9.5 | May 2001 | Aug 2001 | |
NIST | - | - | Aug 2001 | Sep 2001 | |
NMIJ | 516.68 | * | Sep 2001 | Feb 2003 | |
NIM | 518.88 | 5.2 | Feb 2002 | Mar 2002 | |
KRISS | 528.41 | 5.9 | Mar 2002 | Nov 2002 | |
CMI | 520.73 | 2.5 | May 2002 | Sep 2002 | |
GUM | 527.90 | 11.0 | Sep 2002 | Oct 2002 | |
NPL | 524.86 | 1.6 | Oct 2002 | Feb 2003 | |
PTB | 522.89 | 4.6 | Feb 2003 | Mar 2003 | |
'- : no measurements available | * : no official uncertainty value |
COOMET.M.H-K1.c
The measurement results are listed in Table 1 of the COOMET.M.H-K1.c Final Report.
COOMET.M.H-K1
The measurement results are listed in Table 7 of the COOMET.M.H-K1 Final Report.
APMP.M.H-K1.c
The measurement results are listed in Table 7.2 of the APMP.M.H-K1.c Final Report.
EURAMET.M.H-K1.c
The measurement results are listed in Table 5 of the EURAMET.M.H-K1.c Final Report.
CCM.H-K1.c, COOMET.M.H-K1.c, COOMET.M.H-K1, APMP.M.H-K1.c and EURAMET.M.H-K1.c
Hardness Vickers Scale HV30
MEASURAND: Hardness
NOMINAL VALUE: 840 HV
CCM.H-K1.c
x_{i} result obtained by laboratory i expressed as the mean value of 8 repetiton measurements
u_{i} combined standard uncertainty of x_{i}
Lab_{i} | x_{i} | u_{i} | Reception | Results available | |
of package | at Pilot Laboratory | ||||
INRIM | 820.85 | 8.6 | Mar 2001 | May 2001 | |
INMETRO | 841.55 | * | May 2001 | Aug 2001 | |
NIST | - | - | Aug 2001 | Sep 2001 | |
NMIJ | 811.59 | 11.0 | Sep 2001 | Feb 2003 | |
NIM | 815.91 | 10.6 | Feb 2002 | Mar 2002 | |
KRISS | 831.86 | 10.8 | Mar 2002 | Nov 2002 | |
CMI | 810.64 | * | May 2002 | Sep 2002 | |
GUM | 818.03 | 17.0 | Sep 2002 | Oct 2002 | |
NPL | 837.17 | * | Oct 2002 | Feb 2003 | |
PTB | 817.86 | 8.8 | Feb 2003 | Mar 2003 | |
'- : no measurements available | * : no official uncertainty value |
COOMET.M.H-K1.c
The measurement results are listed in Table 1 of the COOMET.M.H-K1.c Final Report.
COOMET.M.H-K1
The measurement results are listed in Table 7 of the COOMET.M.H-K1 Final Report.
APMP.M.H-K1.c
The measurement results are listed in Table 7.3 of the APMP.M.H-K1.c Final Report.
EURAMET.M.H-K1.c
The measurement results are listed in Table 5 of the EURAMET.M.H-K1.c Final Report.