CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Zinc freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Tin freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Indium freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Gallium melting point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Mercury triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Argon triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Zinc freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
INMETRO | -0.69 | 1.37 | ||
INRIM | -0.61 | 0.91 | ||
INTI | -1.56 | 3.37 | ||
KRISS | 0.61 | 0.99 | ||
LNE-LCM/Cnam | 1.01 | 0.96 | ||
NIM | 0.13 | 0.74 | ||
NIST | 0.02 | 0.61 | ||
NMIA | -1.07 | 0.64 | ||
NMIJ/AIST | 0.19 | 0.78 | ||
NPL | -1.78 | 0.64 | ||
NRC | 0.30 | 0.75 | ||
PTB | 1.69 | 1.23 | ||
VSL | 2.53 | 1.04 | ||
TTBS | -1.2 | 11.8 |
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Tin freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
INMETRO | -0.91 | 0.84 | ||
INRIM | 0.09 | 0.59 | ||
INTI | -1.10 | 1.88 | ||
KRISS | 0.22 | 0.75 | ||
LNE-LCM/Cnam | 0.29 | 0.64 | ||
NIM | 0.33 | 0.54 | ||
NIST | -0.11 | 0.37 | ||
NMIA | -0.21 | 0.46 | ||
NMIJ/AIST | -0.03 | 0.53 | ||
NPL | 0.09 | 0.50 | ||
NRC | 0.65 | 0.47 | ||
PTB | 0.28 | 0.79 | ||
VSL | 0.54 | 0.56 | ||
TTBS | -3.0 | 10.8 |
This is the possibility to place a comment.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Indium freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
INMETRO | -0.65 | 0.61 | ||
INRIM | -0.02 | 0.58 | ||
INTI | 0.45 | 2.35 | ||
KRISS | 0.27 | 0.97 | ||
LNE-LCM/Cnam | 0.64 | 0.49 | ||
NIM | 0.13 | 0.43 | ||
NIST | 0.13 | 0.25 | ||
NMIA | -0.04 | 0.39 | ||
NMIJ/AIST | 0.16 | 0.36 | ||
NPL | -0.29 | 0.29 | ||
NRC | 0.12 | 0.29 | ||
PTB | 0.06 | 0.75 | ||
VSL | -0.02 | 0.40 |
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Gallium melting point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
BIPM | 0.11 | 0.32 | ||
INMETRO | 0.29 | 0.34 | ||
INRIM | 0.04 | 0.35 | ||
INTI | 0.15 | 0.87 | ||
KRISS | -0.08 | 0.41 | ||
LNE-LCM/Cnam | -0.36 | 0.38 | ||
NIM | 0.26 | 0.38 | ||
NIST | -0.22 | 0.23 | ||
NMIA | -0.33 | 0.35 | ||
NMIJ/AIST | -0.24 | 0.30 | ||
NPL | -0.06 | 0.26 | ||
NRC | 0.12 | 0.27 | ||
PTB | -0.05 | 0.30 | ||
VSL | -0.14 | 0.36 | ||
TTBS | -1.0 | 6.4 |
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Mercury triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
INMETRO | -0.11 | 0.41 | ||
INRIM | 0.18 | 0.30 | ||
INTI | -0.08 | 0.66 | ||
KRISS | -0.19 | 0.40 | ||
LNE-LCM/Cnam | 0.76 | 0.64 | ||
NIM | 0.24 | 0.31 | ||
NIST | -0.32 | 0.25 | ||
NMIA | -0.04 | 0.29 | ||
NMIJ/AIST | 0.18 | 0.59 | ||
NPL | 0.06 | 0.24 | ||
NRC | 0.01 | 0.24 | ||
PTB | 0.12 | 0.29 | ||
VSL | -0.07 | 0.25 | ||
TTBS | 1.1 | 5.9 |
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Argon triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Degrees of equivalence represented by D_{i} = (T_{I} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence, both expressed in mK.
LAB_{i} | D_{i} | U_{neg,i} | U_{pos,i} | |
---|---|---|---|---|
mK |
mK |
mK |
||
INMETRO | 0.21 | 0.79 | ||
INRIM | 0.22 | 1.08 | ||
INTI | 0.95 | 1.74 | ||
KRISS | 0.28 | 1.03 | ||
LNE-LCM/Cnam | -0.17 | 0.80 | ||
NIM | -0.67 | 0.66 | ||
NIST | -1.41 | 0.54 | ||
NMIA | -0.36 | 0.99 | ||
NMIJ/AIST | 0.20 | 1.30 | ||
NPL | 0.35 | 0.66 | ||
NRC | -0.06 | 0.65 | ||
PTB | 0.10 | 0.73 | ||
VSL | -0.82 | 0.63 |
Metrology area, Sub-field | Thermometry, Standard Platinum Resistance Thermometers |
Description | Realizations of the ITS-90 from 234.315 6 K to 933.473 K |
Time of measurements | 2021 - 2022 |
Status | Report in progress, draft B |
References | |
Measurand | Temperature: from the triple point of Hg (234.315 6 K) to the freezing point of Al (933.473 K) |
Transfer device | Standard Platinum Resistance Thermometers |
Comparison type | Key Comparison |
Consultative Committee | CCT (Consultative Committee for Thermometry) |
Conducted by | CCT (Consultative Committee for Thermometry) |
Pilot institute |
CEM
Centro Español de Metrologia Spain |
Contact person | Raúl Caballero +34918074822 |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCT-K9 and SIM.T-K9.3
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Zinc freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
CCT-K9
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = -0.20 mK u_{R} = 0.11 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
Linking SIM.T-K9.3 to CCT-K9
The INMETRO provides the link between the key comparison SIM.T-K9.3 to CCT-K9, having participated in both comparisons. The linking process is described in Section 7 of the the SIM.T-K9.3 Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Tin freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = 0.06 mK u_{R} = 0.08 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
Linking SIM.T-K9.3 to CCT-K9
The INMETRO provides the link between the key comparison SIM.T-K9.3 to CCT-K9, having participated in both comparisons. The linking process is described in Section 7 of the the SIM.T-K9.3 Final Report.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Indium freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = -0.08 mK u_{R} = 0.05 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Gallium melting point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = 0.23 mK u_{R} = 0.03 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
Linking SIM.T-K9.3 to CCT-K9
The INMETRO provides the link between the key comparison SIM.T-K9.3 to CCT-K9, having participated in both comparisons. The linking process is described in Section 7 of the the SIM.T-K9.3 Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Mercury triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = 0.28 mK u_{R} = 0.04 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
Linking SIM.T-K9.3 to CCT-K9
The INMETRO provides the link between the key comparison SIM.T-K9.3 to CCT-K9, having participated in both comparisons. The linking process is described in Section 7 of the the SIM.T-K9.3 Final Report.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Argon triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
Key comparison reference value:
Resistance ratios are used with calibration equations for each thermometer to obtain temperature T_{i}.
The key comparison reference temperature, T_{R}, is calculated from the mean of the average temperatures from the participants' results, excluded SRPT measurements that fail the cutoff criteria described in section 3.5 of the Final Report, and its associated standard uncertainty u_{R} is computed as explain in section 2.4 of the Final Report.
T_{R} = 0.80 mK u_{R} = 0.08 mK
The degree of equivalence of laboratory i relative to the key comparison reference value T_{R} is given by a pair of terms, both expressed in mK:
D_{i} = (T_{i} - T_{R}) and its expanded uncertainty U_{i} at a 95 % level of confidence is computed as described in section 2.3 of the Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Zinc freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.
SIM.T-K9.3
The measurement results are listed in Table 3 and 4 of the SIM.T-K9.3 Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Tin freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.
SIM.T-K9.3
The measurement results are listed in Table 3 and 4 of the SIM.T-K9.3 Final Report.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Indium freezing point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Gallium melting point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.
SIM.T-K9.3
The measurement results are listed in Table 3 and 4 of the SIM.T-K9.3 Final Report.
CCT-K9 and SIM.T-K9.3
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Mercury triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.
SIM.T-K9.3
The measurement results are listed in Table 3 and 4 of the SIM.T-K9.3 Final Report.
CCT-K9
MEASURAND Resistance Ratio, W, at nominal temperature
NOMINAL TEMPERATURE Argon triple point, a defined fixed point on the International Temperature Scale of 1990 (ITS-90)
W_{i} dimensionless resistance ratio measurement results of thermometer from laboratory i
u_{i} combined standard uncertainty of measurement made using thermometer from laboratory i are expressed in mK
The laboratory measurement results are listed in Appendix B of the CCT-K9 Final Report.