CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
U_{IOCSIC} = 10.62 %
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
Red diamonds: participants in CCPRK2.b
Blue triangles: participants in EURAMET.PRK2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
Degrees of equivalence relative to the CCPRK2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{2}
In dark blue: participants in CCPRK2.b
In blue: participants in EURAMET.PRK2.b only
Linkage is ensured by the common participation of LNEINM, IOCSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PRK2.b results (August 2013).
The full matrix of equivalence is available from CCPRK2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
Metrology area, Subfield  Photometry and Radiometry, Radiometry 
Description  Spectral responsivity 
Time of measurements  2003  2004 
Status  Approved for equivalence 
References  
Measurand  Spectral responsivity 
Parameters  Wavelength from 300 nm to 1000 nm 
Transfer device  Si traps and photodiodes 
Comparison type  Key Comparison 
Consultative Committee  CCPR (Consultative Committee for Photometry and Radiometry) 
Conducted by  EURAMET (European Association of National Metrology Institutes) 
RMO Internal Identifier  EUROMET Project No 587 
Comments 
Results published on 13 October 2013 EURAMET.PRK2.b results are linked to those of CCPRK2.b Spectral responsivity

Pilot institute 
IOCSIC
Instituto de Optica Daza de Valdés Spain 
Contact person  Joaquin CamposAcosta joaquin.campos@csic.es 915616800 
Pilot laboratory  

IOCSIC 
Instituto de Optica Daza de Valdés, Spain, EURAMET 
CMI 
Czech Metrology Institute, Czechia, EURAMET 
GUM 
Glowny Urzad Miar, Central Office of Measures, Poland, EURAMET 
INM 
National Institute of Metrology, Romania, EURAMET 
JV 
Justervesenet, Norway, EURAMET 
LNEINM 
Conservatoire national des arts et métiers / Institut National de Métrologie, France, EURAMET 
METAS 
Federal Institute of Metrology, Switzerland, EURAMET 
SP 
Technical Research Institute of Sweden from 2017 Research Institutes of Sweden AB, Sweden, EURAMET 
UME 
TÜBITAK Ulusal Metroloji Enstitüsü, Turkey, EURAMET 
VSL 
VSL, Netherlands, EURAMET 
This page proposes printout on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPRK2.b Final report). 
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: 
The degree of equivalence between two laboratories is given by a pair of terms: 
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPRK2.b Final Report.
The EURAMET.PRK2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPRK2.b key comparison reference value is established through the common participation of LNEINM, IOCSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPRK2.b to the laboratories having participated in EURAMET.PRK2.b only.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.
CCPRK2.b and EURAMET.PRK2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
The CCPRK2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPRK2.b Final Report for all wavelengths.
The EURAMET.PRK2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PRK2.b Final Report.