CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
U_{IO-CSIC} = 10.62 %
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
Red diamonds: participants in CCPR-K2.b
Blue triangles: participants in EURAMET.PR-K2.b only
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
Degrees of equivalence relative to the CCPR-K2.b key comparison reference value:
D_{i} and U_{i} (k = 2) expressed in 10^{-2}
In dark blue: participants in CCPR-K2.b
In blue: participants in EURAMET.PR-K2.b only
Linkage is ensured by the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons.
The acronyms given here are those in use at the time of publication of EURAMET.PR-K2.b results (August 2013).
The full matrix of equivalence is available from CCPR-K2.b Results (PDF file).
Linked results are presented under A4 printable format in Summary Results (PDF file dated August 2013).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO-31-0 Annex B Rule B.
Metrology area, Sub-field | Photometry and Radiometry, Properties of detectors and sources |
Description | Spectral responsivity |
Time of measurements | 2000 - 2001 |
Status | Approved for equivalence |
Final Reports of the comparisons | |
References | |
Measurand | Spectral responsivity |
Parameters | Wavelength from 300 nm to 1000 nm (18 values) |
Transfer device | Si traps and photodiodes |
Comparison type | Key Comparison |
Consultative Committee | CCPR (Consultative Committee for Photometry and Radiometry) |
Conducted by | CCPR (Consultative Committee for Photometry and Radiometry) |
Comments | Results published on 21 October 2004 EURAMET.PR-K2.b results are linked to those of CCPR-K2.b Spectral responsivity |
Pilot institute |
BIPM
Bureau International des Poids et Mesures BIPM - International Organization |
Contact person | R. Goebel +33 (0) 1 45 07 70 69 |
Pilot laboratory | |
---|---|
BIPM |
Bureau International des Poids et Mesures, BIPM - International Organization, N/A |
KRISS |
Korea Research Institute of Standards and Science, Korea, Republic of, APMP |
MSL |
Measurement Standards Laboratory, New Zealand, APMP |
NIST |
National Institute of Standards and Technology, United States, SIM |
NMIJ AIST |
National Metrology Institute of Japan, Japan, APMP |
NPL |
National Physical Laboratory, United Kingdom, EURAMET |
NRC |
National Research Council, Canada, SIM |
PTB |
Physikalisch-Technische Bundesanstalt, Germany, EURAMET |
SMU |
Slovensky Metrologicky Ustav, Slovakia, EURAMET |
VNIIOFI |
All-Russian Scientific Institute for Optical and Physical Measurements, Rosstandart, Russian Federation, COOMET |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
The key comparison reference value, x_{R}, is calculated as the weighted mean of the participant's results, x_{i}, weighted by the inverse square of the individual uncertainties, u_{i}, with weights limited to a maximum of 20% (see details in Section 4.2 page 58 of the CCPR-K2.b Final report). |
The degree of equivalence of the each laboratory with respect to the key comparison reference value is given by a pair of terms: |
The degree of equivalence between two laboratories is given by a pair of terms: |
The detailed computation of the expanded uncertainties U_{i} and U_{ij} is given on pages 62 and 63 of the CCPR-K2.b Final Report.
The EURAMET.PR-K2.b individual results are first treated inside EURAMET through the computation of a EURAMET reference value and corresponding degrees of equivalence (see Section 7 of the Final Report).
Linkage to the CCPR-K2.b key comparison reference value is established through the common participation of LNE-INM, IO-CSIC and VSL in both key comparisons (see Section 8 of the Final Report).
This makes it possible to extend the table of degrees of equivalence and the graph of equivalence obtained for CCPR-K2.b to the laboratories having participated in EURAMET.PR-K2.b only.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 300 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 320 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 340 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 360 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 380 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 400 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 450 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 500 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 550 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 600 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 650 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 700 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 750 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 800 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 850 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 900 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 950 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.
CCPR-K2.b and EURAMET.PR-K2.b
MEASURAND : spectral responsivity
WAVELENGTH : 1000 nm
The CCPR-K2.b laboratory individual measurements correspond to calibrations, expressed in A/W, of the BIPM reference by transfer of detectors.
These values, x_{i}, together with the associated relative standard uncertainties, u_{i}, are given in Table 48, page 64, of the CCPR-K2.b Final Report for all wavelengths.
The EURAMET.PR-K2.b individual laboratory measurements are given in Section 5 and 6, starting page 14, of the EURAMET.PR-K2.b Final Report.