CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
Green triangle : EUROMET.EM.RFK8.1.CL participant
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
Green triangle : EUROMET.EM.RFK8.1.CL participant
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
Green triangle : EUROMET.EM.RFK8.1.CL participant
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Graph of equivalence : D_{i} and expanded uncertainty U_{i}, both dimensionless
Click on the graph for a closer view
Red diamonds : CCEM.RFK8.CL participants
Blue squares : EUROMET.EM.RFK8.CL participants
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
 : no measurements
* : TM3 was not measured
In blue : participants in EUROMET.EM.RFK8.CL
In green : participant in EUROMET.EM.RFK8.1.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
 : no measurements at this frequency
N/A : not applicable (no measurements at all on the specified device)
In blue : participants in EUROMET.EM.RFK8.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
 : no measurements
* : TM3 was not measured
In blue : participants in EUROMET.EM.RFK8.CL
In green : participant in EUROMET.EM.RFK8.1.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
N/A : not applicable (no measurements at all on the specified device)
In blue : participants in EUROMET.EM.RFK8.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
 : no measurements
* : TM3 was not measured
In blue : participants in EUROMET.EM.RFK8.CL
In green : participant in EUROMET.EM.RFK8.1.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
Degrees of equivalence, D_{i} and expanded uncertainty U_{i}, both dimensionless
 : no measurements at this frequency
N/A : not applicable (no measurements at all on the specified device)
In blue : participants in EUROMET.EM.RFK8.CL
Results are presented under A4 printable format in Summary Results (.PDF file).
Unless otherwise stated, in the final numbers presented here, rounding has been applied according to ISO310 Annex B Rule B.
Metrology area, Subfield  Electricity and Magnetism, Radio frequencies 
Description  Power in 50 ohm coaxial lines 
Time of measurements  2001  2002 
Status  Approved for equivalence 
Measurand  Calibration factor in coaxial 7 mm transmission line 
Parameters  Frequency: 10 MHz to 18 GHz (8 discrete values) 
Transfer device  Three thermistor mounts identified as TM4, TM5, and TM6 TM4 and TM6 have a male TypeN 50 ohm connector, and TM5 has a GPC7connector 
Comparison type  Key Comparison 
Consultative Committee  CCEM (Consultative Committee for Electricity and Magnetism) 
Conducted by  EURAMET (European Association of National Metrology Institutes) 
RMO Internal Identifier  EUROMET Project No 633 
Comments  EUROMET.EM.RFK8.CL and EUROMET.EM.RFK8.1.CL results are linked to those of CCEM.RFK8.CL CCEM.RFK8.CL results published on 01 June 2005

Pilot institute 
VSL
VSL Netherlands 
Contact person  J.P.M. de Vreede +31 (0) 15 269 1621 
INRIM 
Istituto Nazionale di Ricerca Metrologica, Italy, EURAMET 
METAS 
Federal Institute of Metrology, Switzerland, EURAMET 
NMiVSL 
Nederlands Meetinstituut  Van Swinden Laboratorium (became VSL in 2009), Netherlands, EURAMET 
NPL 
National Physical Laboratory, United Kingdom, EURAMET 
NRC 
National Research Council, Canada, SIM 
SMU 
Slovensky Metrologicky Ustav, Slovakia, EURAMET 
SP 
Technical Research Institute of Sweden from 2017 Research Institutes of Sweden AB, Sweden, EURAMET 
This page proposes printout on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARDS : thermistor mounts of typeN 50 ohm connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM1 at 10 MHz, x_{R} = 0.9681 and u_{R} = 0.0018 
For each travelling standard of typeN connector (TM1 and TM3), the degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{TMn i} = the difference between laboratory i result and x_{R}, and U_{TM}_{n} _{i}, its expanded uncertainty taking into account u_{R} (n = 1 or 3). These are combined to obtain the degree of equivalence of laboratory i for typeN connector, using the following expressions: D_{i} = avg(D_{TM1 i}, D_{TM3 i}) and U_{i} = [[avg(U_{TM1 i}, U_{TM3} _{i})]^{2} + [abs(D_{TM1 i } D_{TM3} _{i})/2]^{2}]^{1/2}. 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NPL, INRIM, NRC, and NMiVSL) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 5 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
The NMiVSL is the pilot laboratory of the bilateral comparison EUROMET.EM.RFK8.1.CL by which the results of MIKES participant in EUROMET.EM.RFK8.1.CL are linked to those obtained in EUROMET.EM.RFK8.CL (see EUROMET.EM.RFK8.1.CL Final Report).
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results of MIKES obtained in EUROMET.EM.RFK8.1.CL.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARDS : thermistor mounts with PC7connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM2 at 10 MHz, x_{R} = 0.9645 and u_{R} = 0.0023 
The degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{i} = the difference between laboratory i result and x_{R}, and U_{i}, its expanded uncertainty taking into account u_{R}. 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NMiVSL and NRC) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 6 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARDS : thermistor mounts of typeN 50 ohm connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM1 at 4 GHz, x_{R} = 0.9818 and u_{R} = 0.0010 
For each travelling standard of typeN connector (TM1 and TM3), the degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{TMn i} = the difference between laboratory i result and x_{R}, and U_{TM}_{n} _{i}, its expanded uncertainty taking into account u_{R} (n = 1 or 3). These are combined to obtain the degree of equivalence of laboratory i for typeN connector, using the following expressions: D_{i} = avg(D_{TM1 i}, D_{TM3 i}) and U_{i} = [[avg(U_{TM1 i}, U_{TM3} _{i})]^{2} + [abs(D_{TM1 i}  D_{TM3} _{i})/2]^{2}]^{1/2}. 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NPL, INRIM, NRC, and NMiVSL) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 5 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
The NMiVSL is the pilot laboratory of the bilateral comparison EUROMET.EM.RFK8.1.CL by which the results of MIKES participant in EUROMET.EM.RFK8.1.CL are linked to those obtained in EUROMET.EM.RFK8.CL (see EUROMET.EM.RFK8.1.CL Final Report).
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results of MIKES obtained in EUROMET.EM.RFK8.1.CL.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARDS : thermistor mounts with PC7connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM2 at 4 GHz, x_{R} = 0.9829 and u_{R} = 0.0011 
The degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{i} = the difference between laboratory i result and x_{R}, and U_{i}, its expanded uncertainty taking into account u_{R}. 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NMiVSL and NRC) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 6 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARDS : thermistor mounts of typeN 50 ohm connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM1 at 18 GHz, x_{R} = 0.9391 and u_{R} = 0.0021 
For each travelling standard of typeN connector (TM1 and TM3), the degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{TMn i} = the difference between laboratory i result and x_{R}, and U_{TM}_{n} _{i}, its expanded uncertainty taking into account u_{R} (n = 1 or 3). These are combined to obtain the degree of equivalence of laboratory i for typeN connector, using the following expressions: D_{i} = avg(D_{TM1 i}, D_{TM3 i}) and U_{i}_{ }= [[avg(U_{TM1 i}, U_{TM3} _{i})]^{2} + [abs(D_{TM1 i}  D_{TM3} _{i})/2]^{2}]^{1/2} 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NPL, INRIM, NRC, and NMiVSL) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 5 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
The NMiVSL is the pilot laboratory of the bilateral comparison EUROMET.EM.RFK8.1.CL by which the results of MIKES participant in EUROMET.EM.RFK8.1.CL are linked to those obtained in EUROMET.EM.RFK8.CL (see EUROMET.EM.RFK8.1.CL Final Report).
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results of MIKES obtained in EUROMET.EM.RFK8.1.CL.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARDS : thermistor mounts with PC7connector
For each travelling standard, the key comparison reference value, x_{R}, is obtained as the unweighted average of the results of a number of selected laboratories (see section 7.2 on page 17 of the CCEM.RFK8.CL Final Report). The standard uncertainty, u_{R}, of the key comparison reference value is the standard deviation of the mean. x_{R} and u_{R} are dimensionless quantities. For TM2 at 18 GHz, x_{R} = 0.9434 and u_{R} = 0.0017 
The degree of equivalence of laboratory i with respect to the reference value is given by a pair of terms, both dimensionless: D_{i} = the difference between laboratory i result and x_{R}, and U_{i}, its expanded uncertainty taking into account u_{R}. 
The degree of equivalence between two laboratories i and j is given by a pair of terms, both dimensionless: D_{ij} = (D_{i}  D_{j}) and U_{ij}, its expanded uncertainty (k = 2), U_{ij} = (U_{i}^{2} + U_{j}^{2})^{1/2}_{.}The pairwise degrees of equivalence are not explicitely calculated. 
The laboratories that are participants in both comparisons and which have an independent realisation of the quantity power (NMiVSL and NRC) are used for linking the results obtained in EUROMET.EM.RFK8.CL to the results obtained in CCEM.RFK8.CL. The linking process is explained in section 6 of the Linkage Report.
It follows that the table of degrees of equivalence and the graph of equivalence obtained for CCEM.RFK8.CL are extended with results obtained in EUROMET.EM.RFK8.CL.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 10 MHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 4 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.
CCEM.RFK8.CL, EUROMET.EM.RFK8.CL and
EUROMET.EM.RFK8.1.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARDS FOR CCEM.RFK8.CL : TM1 and TM3 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.CL : TM4 and TM6 both having a male typeN 50 ohm connector
TRAVELLING STANDARDS FOR EUROMET.EM.RFK8.1.CL : DUT1 and DUT2 both having a male typeN 50 ohm connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.
CCEM.RFK8.CL and EUROMET.EM.RFK8.CL
MEASURAND : Calibration factor in coaxial 7 mm transmission line
NOMINAL VALUE : 1.00
FREQUENCY : 18 GHz
TRAVELLING STANDARD FOR CCEM.RFK8.CL : TM2 with a PC7connector
TRAVELLING STANDARD FOR EUROMET.EM.RFK8.CL : TM5 with a PC7connector
The actual calibration factors of the devices under test are not relevant for the quality of the measurement results, which are given here as the difference between the laboratory result and the relevant key comparison reference value.
Please see Equivalence statements.