CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Brown squares: participants in EUROMET.EM-K11
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 20 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Brown squares: participants in EUROMET.EM-K11
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 100 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Brown squares: participants in EUROMET.EM-K11
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 MHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Brown squares: participants in EUROMET.EM-K11
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Blue circles: participants in SIM.EM-K11 only
Brown squares: participants in EUROMET.EM-K11 only
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 20 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Blue circles: participants in SIM.EM-K11 only
Brown squares: participants in EUROMET.EM-K11 only
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 100 kHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Blue circles: participants in SIM.EM-K11 only
Brown squares: participants in EUROMET.EM-K11 only
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 MHz
Degrees of equivalence: D_{i} and expanded uncertainty U_{i} (k = 2), both expressed in µV/V
Red diamonds: participants in CCEM-K11
Green triangle: participant in CCEM-K11.1 only
Blue circles: participants in SIM.EM-K11 only
Brown squares: participants in EUROMET.EM-K11 only
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -6 | 40 |
NPL | -4 | 29 |
VSL | -9 | 39 |
NMIA | 0 | 11 |
INTI | 13 | 25 |
NRC | 8 | 35 |
A*STAR | -1 | 33 |
NIM | -41 | 78 |
SP | -2 | 20 |
NIST | 6 | 38 |
JV | 5 | 31 |
INRIM | -17 | 26 |
BEV | 6 | 38 |
MKEH | 1 | 26 |
IPQ | 2 | 61 |
CEM | -5 | 45 |
MIRS/SIQ/Metrology | 0 | 74 |
MIKES | 13 | 33 |
Trescal | -16 | 46 |
LNE | 2 | 23 |
METAS | -30 | 38 |
UME | 5 | 51 |
NMISA | -6 | 62 |
CMI | 2 | 58 |
EIM | 4 | 88 |
NPLI | -7 | 26 |
VNIIM | 2 | 29 |
Green: participant in CCEM-K11.1 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 20 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -13 | 41 |
NPL | -4 | 29 |
VSL | -7 | 39 |
NMIA | 2 | 14 |
INTI | 11 | 28 |
NRC | 2 | 31 |
A*STAR | -2 | 36 |
NIM | -28 | 78 |
SP | -1 | 21 |
NIST | 25 | 40 |
JV | 11 | 32 |
INRIM | -18 | 25 |
BEV | 0 | 38 |
MKEH | -3 | 26 |
IPQ | -6 | 60 |
CEM | -4 | 45 |
MIRS/SIQ/Metrology | -12 | 73 |
MIKES | 8 | 32 |
Trescal | -26 | 46 |
LNE | 6 | 26 |
METAS | -42 | 37 |
UME | 7 | 51 |
NMISA | 17 | 60 |
CMI | 1 | 62 |
EIM | -3 | 83 |
NPLI | -8 | 25 |
VNIIM | 20 | 30 |
Green: participant in CCEM-K11.1 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 100 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -9 | 43 |
NPL | 4 | 40 |
VSL | -5 | 48 |
NMIA | 7 | 39 |
INTI | 13 | 44 |
NRC | 7 | 59 |
A*STAR | -9 | 42 |
NIM | 1 | 95 |
SP | -3 | 30 |
NIST | -3 | 53 |
JV | 19 | 73 |
INRIM | -32 | 45 |
BEV | -3 | 46 |
MKEH | -9 | 37 |
IPQ | -10 | 71 |
CEM | -2 | 65 |
MIRS/SIQ/Metrology | -10 | 123 |
MIKES | -1 | 45 |
Trescal | -32 | 67 |
LNE | -5 | 33 |
METAS | -44 | 58 |
UME | -19 | 60 |
NMISA | 2 | 139 |
CMI | 1 | 84 |
EIM | 39 | 90 |
NPLI | -22 | 30 |
VNIIM | 62 | 69 |
Green: participant in CCEM-K11.1 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 MHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -44 | 100 |
NPL | -46 | 281 |
VSL | 26 | 298 |
NMIA | 3 | 102 |
INTI | -29 | 105 |
NRC | 42 | 84 |
A*STAR | -42 | 369 |
NIM | 138 | 184 |
SP | -24 | 94 |
NIST | 19 | 151 |
JV | 8 | 269 |
INRIM | -141 | 142 |
BEV | -49 | 148 |
MKEH | -172 | 233 |
IPQ | 79 | 287 |
CEM | 2 | 168 |
MIRS/SIQ/Metrology | 31 | 754 |
MIKES | -38 | 147 |
Trescal | -126 | 207 |
LNE | - | - |
METAS | -151 | 129 |
UME | -51 | 132 |
NMISA | -579 | 758 |
CMI | 40 | 243 |
EIM | 126 | 190 |
NPLI | -2 | 83 |
VNIIM | 53 | 178 |
Green: participant in CCEM-K11.1 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | 0.3 | 2.5 |
NPL | -5.1 | 8.9 |
VSL | -1.7 | 7.9 |
NMIA | -0.1 | 5.7 |
INTI | -0.1 | 9.9 |
NRC | 0.7 | 4.8 |
A*STAR | -3.3 | 18.0 |
NIM | 3.7 | 13.9 |
SP | 0.5 | 6.5 |
NIST | -0.9 | 12.3 |
UTE | 5.2 | 18.1 |
INMETRO | -0.8 | 12.8 |
CENAM | -1.3 | 25.1 |
JV | 0.8 | 14.3 |
INRIM | -10.0 | 10.0 |
BEV | 2.4 | 12.7 |
MKEH | 1.6 | 8.0 |
IPQ | 9.5 | 41.2 |
CEM | 3.4 | 10.8 |
MIRS/SIQ/Metrology | 4.4 | 30.3 |
MIKES | 1.4 | 6.2 |
Trescal | -8.7 | 8.5 |
LNE | 2.6 | 7.5 |
METAS | -11.1 | 13.3 |
UME | 1.1 | 13.6 |
NMISA | 6.9 | 18.0 |
CMI | 2.3 | 19.4 |
EIM | 1.6 | 5.6 |
NPLI | -7.7 | 11.5 |
VNIIM | 1.4 | 10.4 |
Green: participant in CCEM-K11.1 only
Blue: participants in SIM.EM-K11 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 20 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -0.3 | 4.3 |
NPL | -3.2 | 8.7 |
VSL | 0.9 | 11.8 |
NMIA | 2.9 | 6.7 |
INTI | 0.9 | 9.8 |
NRC | -0.2 | 5.6 |
A*STAR | -4.2 | 23.9 |
NIM | -3.2 | 15.8 |
SP | 0.3 | 6.5 |
NIST | -2.8 | 12.7 |
UTE | -1.1 | 18.2 |
INMETRO | -1.1 | 12.5 |
CENAM | -5.1 | 24.9 |
JV | 0.3 | 15.6 |
INRIM | -8.4 | 10.4 |
BEV | 0.4 | 14.1 |
MKEH | 0.5 | 9.9 |
IPQ | 6.1 | 48.3 |
CEM | -7.9 | 11.4 |
MIRS/SIQ/Metrology | 1.1 | 30.5 |
MIKES | 0.2 | 7.8 |
Trescal | -9.0 | 9.1 |
LNE | 0.5 | 9.1 |
METAS | -12.1 | 10.0 |
UME | 5.0 | 15.1 |
NMISA | 6.5 | 18.7 |
CMI | 3.4 | 21.7 |
EIM | 2.8 | 6.6 |
NPLI | -1.8 | 12.2 |
VNIIM | 18.2 | 14.1 |
Green: participant in CCEM-K11.1 only
Blue: participants in SIM.EM-K11 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 100 kHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -1.5 | 6.0 |
NPL | -2.6 | 12.7 |
VSL | -0.6 | 14.7 |
NMIA | -1.1 | 15.8 |
INTI | -0.2 | 20.8 |
NRC | 3.5 | 6.6 |
A*STAR | -5.8 | 26.9 |
NIM | 8.1 | 29.9 |
SP | -1.6 | 12.8 |
NIST | 2.7 | 23.1 |
UTE | -2.5 | 38.0 |
INMETRO | -2.5 | 17.8 |
CENAM | 1.1 | 26.0 |
JV | 1.9 | 28.7 |
INRIM | -19.2 | 13.7 |
BEV | -2.6 | 18.9 |
MKEH | -5.3 | 18.7 |
IPQ | 6.6 | 61.5 |
CEM | -11.4 | 31.1 |
MIRS/SIQ/Metrology | -7.5 | 50.7 |
MIKES | -2.7 | 10.2 |
Trescal | -14.6 | 14.5 |
LNE | -4.5 | 15.4 |
METAS | -16.0 | 15.4 |
UME | -8.7 | 18.0 |
NMISA | 9.9 | 35.8 |
CMI | -4.2 | 30.1 |
EIM | -4.9 | 9.4 |
NPLI | -19.4 | 14.4 |
VNIIM | 66.6 | 34.0 |
Green: participant in CCEM-K11.1 only
Blue: participants in SIM.EM-K11 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 MHz
Lab i | D_{i} | U_{i} |
/ (µV/V) | / (µV/V) | |
PTB | -28 | 50 |
NPL | -18 | 67 |
VSL | -1 | 101 |
NMIA | 13 | 77 |
INTI | -30 | 62 |
NRC | 13 | 32 |
A*STAR | -6 | 88 |
NIM | 72 | 99 |
SP | 7 | 63 |
NIST | 1 | 77 |
UTE | -34 | 111 |
INMETRO | 19 | 105 |
CENAM | 0 | 57 |
JV | 11 | 111 |
INRIM | -59 | 71 |
BEV | -29 | 76 |
MKEH | -44 | 155 |
IPQ | -37 | 92 |
CEM | -47 | 92 |
MIRS/SIQ/Metrology | -56 | 403 |
MIKES | -31 | 56 |
Trescal | -58 | 62 |
LNE | - | - |
METAS | -57 | 74 |
UME | -23 | 61 |
NMISA | 194 | 193 |
CMI | -57 | 92 |
EIM | -40 | 50 |
NPLI | 106 | 58 |
VNIIM | 229 | 127 |
Green: participant in CCEM-K11.1 only
Blue: participants in SIM.EM-K11 only
Brown: participants in EUROMET.EM-K11 only
Results are presented under A4 printable format in Summary Results (PDF file).
Metrology area, Sub-field | Electricity and Magnetism, AC Voltage, Current, Power, and AC/DC Transfer |
Description | AC/DC voltage transfer difference at low voltages |
Time of measurements | 2004 - 2005 |
Status | Approved for equivalence |
Final Reports of the comparisons | |
Measurand | AC/DC transfer difference |
Parameters | Voltage: 10 mV and 100 mV Frequency: 1 kHz to 1 MHz |
Transfer device | AC/DC transfer standard Fluke 792A |
Comparison type | Key Comparison |
Consultative Committee | CCEM (Consultative Committee for Electricity and Magnetism) |
Conducted by | CCEM (Consultative Committee for Electricity and Magnetism) |
Comments | This is a bilateral follow-up of CCEM-K11. The same travelling standard and technical protocol are used. CCEM-K11 and CCEM-K11.1 results published on 23 May 2007 EUROMET.EM-K11 results and SIM.EM-K11 results are linked to those of CCEM-K11 and CCEM-K11.1 AC/DC voltage transfer standards |
Pilot institute |
RISE
Research Institutes of Sweden AB Sweden |
Contact person | Karl-Erik Rydler karl-erik.rydler@ri.se +46 105165401 |
This page proposes print-out on A4 paper (portrait) of the comparison details (best printed out using a black and white printer).
Please, select items to be printed out, then click on "OK" :
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = 1.6 µV/V; 2u_{R} = 9.2 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 20 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = 1.0 µV/V; 2u_{R} = 10.4 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 100 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = 2.8 µV/V; 2u_{R} = 15.8 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 MHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = 24 µV/V; 2u_{R} = 49 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = -0.5 µV/V; 2u_{R} = 2.3 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by two terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
INTI and NRC provide the link between SIM.EM-K11 and CCEM-K11. The linkage process is described in section 11 of SIM.EM-K11 Final Report. |
The degree of equivalence of laboratory i participant in SIM.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i}_{ }and U_{i}, its expanded uncertainty (k = 2). |
Pair-wive degrees of equivalence inside SIM.EM-K11 are not computed. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 20 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = -0.3 µV/V; 2u_{R} = 2.9 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by two terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
INTI and NRC provide the link between SIM.EM-K11 and CCEM-K11. The linkage process is described in section 11 of SIM.EM-K11 Final Report. |
The degree of equivalence of laboratory i participant in SIM.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i}_{ }and U_{i}, its expanded uncertainty (k = 2). |
Pair-wive degrees of equivalence inside SIM.EM-K11 are not computed. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 100 kHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = 1.6 µV/V; 2u_{R} = 4.5 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by two terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
INTI and NRC provide the link between SIM.EM-K11 and CCEM-K11. The linkage process is described in section 11 of SIM.EM-K11 Final Report. |
The degree of equivalence of laboratory i participant in SIM.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i}_{ }and U_{i}, its expanded uncertainty (k = 2). |
Pair-wive degrees of equivalence inside SIM.EM-K11 are not computed. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 MHz
The key comparison reference value x_{R} is calculated as the weighted mean of the results of all participants in CCEM-K11. The value of the mean of the seven results of SP is used for the computation of x_{R}, see equations 10 and 11 of the Final Report. The standard uncertainty u_{R} of x_{R} is computed as explained in section 6.1.4 of the Final Report. x_{R} = -7 µV/V; 2u_{R} = 27 µV/V |
The degree of equivalence of laboratory i with respect to the key comparison reference value is given by two terms both expressed in µV/V: D_{i} = (x_{i} - x_{R}) and its expanded uncertainty U_{i} (k = 2). The equations for the computation of U_{i}_{ }are given in section 6.1.6 of the Final Report. |
The degree of equivalence between two laboratories i and j is given by two terms both expressed in µV/V: D_{ij} = (x_{i} - x_{j}) and its expanded uncertainty U_{ij} (k = 2). The equations for the computation of U_{ij }are given in section 6.1.7 of the Final Report. |
INTI and NRC provide the link between SIM.EM-K11 and CCEM-K11. The linkage process is described in section 11 of SIM.EM-K11 Final Report. |
The degree of equivalence of laboratory i participant in SIM.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i}_{ }and U_{i}, its expanded uncertainty (k = 2). |
Pair-wive degrees of equivalence inside SIM.EM-K11 are not computed. |
The degrees of equivalence relative to the EUROMET.EM-K11 reference value are reported in Table 19 (page 30) and in Table 20 (page 33) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
SP, PTB and VSL then provide the link between EUROMET.EM-K11 and CCEM-K11. The linkage process is described in Sections 6.1.10, 6.1.11 and 6.1.12 of the EUROMET.EM-K11 Final Report.
The degree of equivalence of laboratory i participant in EUROMET.EM-K11 with respect to the CCEM-K11 key comparison reference value is given by a pair of terms both expressed in µV/V: D_{i} and U_{i} its expanded uncertainty (k = 2). |
Pair-wise degrees of equivalence inside EUROMET.EM-K11 are reported in Appendix 1 of the EUROMET.EM-K11 Final Report. |
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP |
-0.7 |
22.2 | Sep 01 |
PTB | -4.6 | 41.1 | Oct 01 |
NPL | -2.1 | 30.2 | Dec 01 |
VSL | -7.1 | 40.1 | Jan 02 |
SP | -1.2 | 22.2 | Jul 02 |
NMIA | 1.4 | 14.3 | Aug 02 |
INTI | 14.3 | 26.2 | Oct 02 |
SP | 0.9 | 22.2 | Jan 03 |
NRC | 9.6 | 36.1 | Feb 03 |
SP | 1.9 | 22.2 | Jun 03 |
A*STAR | 0.2 | 34.1 | Jul 03 |
NIM | -39.0 | 78.0 | Aug 03 |
SP | 0.8 | 22.2 | Oct 03 |
SP |
-0.7 |
22.2 |
Jun 04 |
NIST |
7.3 |
36.5 |
Aug 04 |
SP |
-0.9 |
22.2 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 20 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) |
U_{Lab i}/ (µV/V) |
Date of measurement |
SP | -0.7 | 22.8 | Sep 01 |
PTB | -11.7 | 42.4 | Oct 01 |
NPL | -2.8 | 30.5 | Dec 01 |
VSL | -5.7 | 40.4 | Jan 02 |
SP | -2.3 | 22.6 | Jul 02 |
NMIA | 3.4 | 16.9 | Aug 02 |
INTI | 12.4 | 29.5 | Oct 02 |
SP | 1.3 | 22.6 | Jan 03 |
NRC | 3.2 | 32.4 | Feb 03 |
SP | 4.3 | 22.6 | Jun 03 |
A*STAR | -0.7 | 37.4 | Jul 03 |
NIM | -26.7 | 78.2 | Aug 03 |
SP | -0.6 | 22.6 | Oct 03 |
SP |
-0.8 |
22.7 |
Jun 04 |
NIST |
25.8 |
38.9 |
Aug 04 |
SP |
-1.1 |
22.8 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 100 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP | 1.7 | 33.3 | Sep 01 |
PTB | -6.2 | 45.2 | Oct 01 |
NPL | 6.4 | 42.2 | Dec 01 |
VSL | -2.2 | 50.2 | Jan 02 |
SP | 0.6 | 33.2 | Jul 02 |
NMIA | 9.7 | 42.2 | Aug 02 |
INTI | 15.5 | 46.2 | Oct 02 |
SP | -2.1 | 33.3 | Jan 03 |
NRC | 9.9 | 61.1 | Feb 03 |
SP | -1.5 | 33.2 | Jun 03 |
A*STAR | -6.6 | 44.2 | Jul 03 |
NIM | 4.2 | 96.1 | Aug 03 |
SP | -1.1 | 33.2 | Oct 03 |
SP |
1.2 |
33.2 |
Jun 04 |
NIST |
0.3 |
50.6 |
Aug 04 |
SP |
1.1 |
33.3 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 10 mV
FREQUENCY : 1 MHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP |
-4.4 |
100.7 | Sep 01 |
PTB | -20.6 | 107.3 | Oct 01 |
NPL | -21.9 | 283.6 | Dec 01 |
VSL | 50.1 | 300.7 | Jan 02 |
SP | 1.1 | 100.1 | Jul 02 |
NMIA | 26.9 | 108.9 | Aug 02 |
INTI | -5.0 | 111.8 | Oct 02 |
SP | 2.5 | 101.8 | Jan 03 |
NRC | 65.3 | 92.2 | Feb 03 |
SP | -2.8 | 100.4 | Jun 03 |
A*STAR | -18.3 | 370.7 | Jul 03 |
NIM | 161.2 | 188.2 | Aug 03 |
SP | 12.1 | 101.5 | Oct 03 |
SP |
-6.8 |
100.5 |
Jun 04 |
NIST |
43.1 |
143.1 |
Aug 04 |
SP |
-1.8 |
101.3 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP | -0.3 | 6.7 | Sep 01 |
PTB | -0.2 | 3.0 | Oct 01 |
NPL | -5.6 | 9.1 | Dec 01 |
VSL | -2.2 | 8.1 | Jan 02 |
SP | 0.4 | 6.7 | Jul 02 |
NMIA | -0.6 | 6.0 | Aug 02 |
INTI | -0.6 | 10.0 | Oct 02 |
SP | 0.6 | 6.7 | Jan 03 |
NRC | 0.2 | 5.1 | Feb 03 |
SP | -0.3 | 6.7 | Jun 03 |
A*STAR | -3.8 | 18.0 | Jul 03 |
NIM | 3.2 | 14.0 | Aug 03 |
SP | -0.5 | 6.7 | Oct 03 |
SP |
0.0 |
6.7 |
Jun 04 |
NIST |
-1.4 |
12.0 |
Aug 04 |
SP |
0.1 |
6.7 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
d_{Lab i-SIM} : result of measurement carried out by laboratory i participant in SIM.EM-K11
U_{Lab }_{i-SIM} : expanded uncertainty (k = 2) of d_{Lab i-SIM}u_{Lab i-SIM }: standard uncertainty (k = 1) of d_{Lab i-SIM}
Lab i | d_{Lab i-SIM} / (µV/V) |
U_{Lab i-SIM} / (µV/V) |
Date of measurement |
INTI |
8.7 |
12.6 |
04 Mar to 16 Apr 04 |
INMETRO | 8.0 | 8.0 | 28 Jun to 28 July 04 |
CENAM |
7.5 |
23 |
26 to 30 Aug 04 |
NRC |
9.5 |
9.5 |
Sep to Oct 04 |
Lab i | d_{Lab i-SIM} / (µV/V) |
u_{Lab i-SIM} / (µV/V) |
Date of measurement |
UTE | 14.0 | 7.5 | 18 May to 22 Jun 04 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 20 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP | -0.2 | 6.9 | Sep 01 |
PTB | -0.6 | 5.0 | Oct 01 |
NPL | -3.5 | 9.1 | Dec 01 |
VSL | 0.6 | 12.1 | Jan 02 |
SP | 0.9 | 6.9 | Jul 02 |
NMIA | 2.6 | 7.1 | Aug 02 |
INTI | 0.6 | 10.1 | Oct 02 |
SP | -0.4 | 6.9 | Jan 03 |
NRC | -0.5 | 6.1 | Feb 03 |
SP | -0.3 | 6.9 | Jun 03 |
A*STAR | -4.5 | 24.0 | Jul 03 |
NIM | -3.5 | 16.0 | Aug 03 |
SP | -0.4 | 6.9 | Oct 03 |
SP |
0.0 |
6.9 |
Jun 04 |
NIST |
-3.1 |
12.4 |
Aug 04 |
SP |
0.3 |
6.9 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
d_{Lab i-SIM} : result of measurement carried out by laboratory i participant in SIM.EM-K11
U_{Lab }_{i-SIM} : expanded uncertainty (k = 2) of d_{Lab i-SIM}u_{Lab i-SIM }: standard uncertainty (k = 1) of d_{Lab i-SIM}
Lab i | d_{Lab i-SIM} / (µV/V) |
U_{Lab i-SIM} / (µV/V) |
Date of measurement |
INTI |
3.0 |
13 |
04 Mar to 16 Apr 04 |
INMETRO | -2.0 | 8.0 | 28 Jun to 28 July 04 |
CENAM |
-6.0 |
23 |
26 to 30 Aug 04 |
NRC |
-1.8 |
8.5 |
Sep to Oct 04 |
Lab i | d_{Lab i-SIM} / (µV/V) |
u_{Lab i-SIM} / (µV/V) |
Date of measurement |
UTE | -2.0 | 7.7 | 18 May to 22 Jun 04 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 100 kHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP | 0.2 | 13.1 | Sep 01 |
PTB | 0.1 | 6.8 | Oct 01 |
NPL | -1.0 | 13.1 | Dec 01 |
VSL | 1.0 | 15.1 | Jan 02 |
SP | 0.6 | 13.1 | Jul 02 |
NMIA | 0.5 | 16.1 | Aug 02 |
INTI | 1.4 | 21.1 | Oct 02 |
SP | 0.1 | 13.2 | Jan 03 |
NRC | 5.1 | 7.3 | Feb 03 |
SP | -1.1 | 13.1 | Jun 03 |
A*STAR | -4.2 | 27.1 | Jul 03 |
NIM | 9.7 | 30.0 | Aug 03 |
SP | -0.4 | 13.2 | Oct 03 |
SP |
0.1 |
13.1 |
Jun 04 |
NIST |
4.3 |
22.7 |
Aug 04 |
SP |
0.6 |
13.2 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
d_{Lab i-SIM} : result of measurement carried out by laboratory i participant in SIM.EM-K11
U_{Lab }_{i-SIM} : expanded uncertainty (k = 2) of d_{Lab i-SIM}u_{Lab i-SIM }: standard uncertainty (k = 1) of d_{Lab i-SIM}
Lab i | d_{Lab i-SIM} / (µV/V) |
U_{Lab i-SIM} / (µV/V) |
Date of measurement |
INTI |
19.0 |
25 |
04 Mar to 16 Apr 04 |
INMETRO | 18.0 | 13.0 | 28 Jun to 28 July 04 |
CENAM |
21.6 |
23 |
26 to 30 Aug 04 |
NRC |
24.1 |
10.3 |
Sep to Oct 04 |
Lab i | d_{Lab i-SIM} / (µV/V) |
u_{Lab i-SIM} / (µV/V) |
Date of measurement |
UTE | 18.0 | 18 | 18 May to 22 Jun 04 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).
CCEM-K11, CCEM-K11.1, SIM.EM-K11 and EUROMET.EM-K11
MEASURAND : AC/DC voltage transfer difference
NOMINAL VOLTAGE : 100 mV
FREQUENCY : 1 MHz
x_{i} : measurement result of laboratory i corrected as explained in section 6 of the Final Report
U_{Lab i} : expanded uncertainty (k = 2) of x_{i}
Lab i | x_{i}/ (µV/V) | U_{Lab i}/ (µV/V) | Date of measurement |
SP | 0.4 | 62.8 | Sep 01 |
PTB | -35.4 | 50.6 | Oct 01 |
NPL | -25.2 | 67.5 | Dec 01 |
VSL | -8.0 | 101.3 | Jan 02 |
SP | -6.4 | 62.2 | Jul 02 |
NMIA | 5.6 | 77.7 | Aug 02 |
INTI | -36.9 | 62.1 | Oct 02 |
SP | 10.3 | 65.2 | Jan 03 |
NRC | 6.4 | 32.9 | Feb 03 |
SP | -1.2 | 62.5 | Jun 03 |
A*STAR | -12.8 | 87.9 | Jul 03 |
NIM | 64.8 | 99.4 | Aug 03 |
SP | 1.1 | 63.6 | Oct 03 |
SP |
-8.0 |
62.8 |
Jun 04 |
NIST |
-6.4 |
71.8 |
Aug 04 |
SP |
3.9 |
64.1 |
Feb 05 |
Laboratory participant in CCEM-K11.1 |
d_{Lab i-SIM} : result of measurement carried out by laboratory i participant in SIM.EM-K11
U_{Lab }_{i-SIM} : expanded uncertainty (k = 2) of d_{Lab i-SIM}u_{Lab i-SIM }: standard uncertainty (k = 1) of d_{Lab i-SIM}
Lab i | d_{Lab i-SIM} / (µV/V) |
U_{Lab i-SIM} / (µV/V) |
Date of measurement |
INTI |
14.0 |
75 |
04 Mar to 16 Apr 04 |
INMETRO | 79.0 | 98.0 | 28 Jun to 28 July 04 |
CENAM |
59.9 |
42 |
26 to 30 Aug 04 |
NRC |
75.0 |
23.0 |
Sep to Oct 04 |
Lab i | d_{Lab i-SIM} / (µV/V) |
u_{Lab i-SIM} / (µV/V) |
Date of measurement |
UTE | 26 | 52 | 18 May to 22 Jun 04 |
The values reported by the participants in EUROMET.EM-K11 are given in Table 9 (page 14) and Table 10 (page 18) of the EUROMET.EM-K11 Final Report for 100 mV and 10 mV, respectively.
Measurements were conducted between June 2005 and Aoril 2008.
The results after correction for the drift of the travelling standard are reported in Section 6.1.8 of the EUROMET.EM-K11 Final Report (page 22 to page 29).