Results for: lithuania
Filters
CMC Quick search Filters close;
?
Added keywords act as an additional filter. Keywords separated by a space will be treated by the logical function AND automatically. You may also use OR. Keyword searches also find close matches. Placing quotation marks (") before and after the search word, e.g. "word", will allow for exact matches.

In case of multiple search, please use ADVANCED SEARCH.
73 results
Reset all
CMC Area
Geographical location
In the CMCs uncertainty statements, the notation Q[a, b] stands for the root-sum-square of the terms between brackets: Q[a, b] = [a 2 + b 2]1/2
Unless otherwise stated the expanded uncertainties given below correspond to k = 2 (at a 95 % level of confidence)
Lithuania, FTMC (Center for Physical Sciences and Technology)
DC voltage (up to 1100 V) , DC voltage sources: single values : 1 V to 10 V
Solid state voltage standard
Relative expanded uncertainty : 5.0E-2 µV/V to 0.3 µV/V
{"row_1":{"col_1":"","col_2":"Expanded uncertainty / (µV/V)","col_3":"Reference Standard used in calibration"},"row_2":{"col_1":"1 V","col_2":"0.06","col_3":"Josephson array voltage standard"},"row_3":{"col_1":"1.018 V","col_2":"0.06","col_3":"Josephson array voltage standard"},"row_4":{"col_1":"10 V","col_2":"0.05","col_3":"Josephson array voltage standard"},"row_5":{"col_1":"1 V","col_2":"0.3","col_3":"Solid state voltage standard"},"row_6":{"col_1":"1.018 V","col_2":"0.3","col_3":"Solid state voltage standard"},"row_7":{"col_1":"10 V","col_2":"0.3","col_3":"Solid state voltage standard"}}
Comparison
Fixed voltage : 1 V, 1.018 V, 10 V
Approved on 06 August 2013
Institute service identifier : FTMC/1
DC resistance , DC resistance standards and sources: low values : 1 Ω
Fixed resistor
Relative expanded uncertainty : 0.3 µΩ/Ω
Direct current comparator bridge
Power : 1 mW
Approved on 06 August 2013
Institute service identifier : FTMC/10
DC resistance , DC resistance standards and sources: intermediate values : 10 kΩ to 1000 kΩ
Fixed resistor
Relative expanded uncertainty : 0.3 µΩ/Ω to 0.6 µΩ/Ω
{"row_1":{"col_1":"","col_2":"Expanded uncertainty"},"row_2":{"col_1":"10 Ω","col_2":"0.3"},"row_3":{"col_1":"100 Ω","col_2":"0.3"},"row_4":{"col_1":"1000 Ω","col_2":"0.4"},"row_5":{"col_1":"10 kΩ, 100 kΩ","col_2":"0.3"},"row_6":{"col_1":"1 MΩ","col_2":"0.6"},"row_7":{"col_1":"10 MΩ","col_2":"1.3"},"row_8":{"col_1":"100 MΩ","col_2":"2.2"},"row_9":{"col_1":"1 GΩ","col_2":"7.1"},"row_10":{"col_1":"10 GΩ","col_2":"120"},"row_11":{"col_1":"100 GΩ","col_2":"990"},"row_12":{"col_1":"1 TΩ","col_2":"9800"}}
The expanded uncertainties given in this table are expressed in µΩ/Ω
Binary voltage divider bridge
Voltage : 5 V
Approved on 06 August 2013
Institute service identifier : FTMC/14
DC voltage (up to 1100 V) , DC voltage meters: intermediate values : 0.1 V to 1000 V
DC voltmeter, multimeter, multifunction transfer standard
Relative expanded uncertainty : 6.8 µV/V to 2.5 µV/V
{"row_1":{"col_1":"","col_2":"Expanded uncertainty"},"row_2":{"col_1":"0.1 V","col_2":"6.8"},"row_3":{"col_1":"1 V","col_2":"3"},"row_4":{"col_1":"10 V","col_2":"1.4"},"row_5":{"col_1":"1000 V","col_2":"2.5"}}
The expanded uncertainties given in this table are expressed in µV/V
Direct measurement
Approved on 06 August 2013
Institute service identifier : FTMC/37
DC resistance , DC resistance meters: low values : 1 Ω
Micro-ohmmeter, multimeter, multifunction transfer standard, resistance bridge
Relative expanded uncertainty : 2 µΩ/Ω
Comparison to standard resistor
Approved on 06 August 2013
Institute service identifier : FTMC/17
DC resistance , DC resistance standards and sources: multiple ranges : 10 MΩ to 100 MΩ
Multifunction calibrator
Relative expanded uncertainty : 3 µΩ/Ω to 6 µΩ/Ω
{"row_1":{"col_1":"","col_2":"Expanded uncertainty"},"row_2":{"col_1":"10 MΩ","col_2":"3"},"row_3":{"col_1":"100 MΩ","col_2":"6"}}
The expanded uncertainties given in this table are expressed in µΩ/Ω
Direct comparison
Approved on 06 August 2013
Institute service identifier : FTMC/32
Items for defining ITS-90 , Temperature : 232 °C to 419 °C
Long-stem Standard Platinum Resistance Thermometer (SPRT)
Absolute expanded uncertainty : 4.0E-3 °C
Calibration at fixed points
Fixed points : H2O, Sn, Zn
Approved on 17 January 2013
Items for defining ITS-90 , Temperature : 29.7646 °C
Long-stem Standard Platinum Resistance Thermometer (SPRT)
Absolute expanded uncertainty : 6.0E-4 °C
Calibration at gallium melting point
Temperature controlled bath
Approved on 20 May 2010
Items for defining ITS-90 , Temperature : 660.323 °C
Long-stem Standard Platinum Resistance Thermometer (SPRT)
Absolute expanded uncertainty : 6.0E-3 °C
Calibration at Aluminium freezing point
3-zone furnace
Approved on 20 May 2010
Frequency , Frequency : 5 MHz
Local frequency standard
Relative expanded uncertainty : 2.0E-13 Hz/Hz
Phase time measurement
Averaging time : 1 day
Approved on 18 November 2009
Institute service identifier : FTMC/7
Various dimensional , Hand Instruments : 0 mm to 1000 mm
Calliper: error of indicated size L
Absolute expanded uncertainty : 6.0E1 µm to 6.1E1 µm
Q[60 µm, 10E-06 L]
Comparison to gauge block
Resolution : 0.05 mm
Approved on 22 March 2005
Institute service identifier : FTMC/6
Time scale difference , Time scale difference : -1 s to 1 s
Local clock vs UTC(FTMC)
Absolute expanded uncertainty : 2 ns
Time interval measurement
Pulse amplitude : > 1 V
Fractional part of modulo 1 second difference, integer part can be provided also
Approved on 18 November 2009
Institute service identifier : FTMC/1
DC resistance , DC resistance meters: intermediate values : 1 MΩ to 1000 MΩ
Ohmmeter, multimeter, multifunction transfer standard, resistance bridge
Relative expanded uncertainty : 2 µΩ/Ω to 9 µΩ/Ω
{"row_1":{"col_1":"","col_2":"Expanded uncertainty"},"row_2":{"col_1":"1 MΩ","col_2":"2"},"row_3":{"col_1":"10 MΩ","col_2":"2"},"row_4":{"col_1":"100 MΩ","col_2":"4"},"row_5":{"col_1":"1 GΩ","col_2":"9"}}
The expanded uncertainties given in this table are expressed in µΩ/Ω
Comparison to standard resistor
Approved on 06 August 2013
Institute service identifier : FTMC/21
DC voltage (up to 1100 V) , DC voltage sources: intermediate values : 10 V to 1000 V
DC voltage source, multifunction calibrator
Relative expanded uncertainty : 1 µV/V to 2 µV/V
Direct measurement
Approved on 06 August 2013
Institute service identifier : FTMC/41
Various dimensional , Hand Instruments : 0 mm to 1 mm
Dial gauge: error of indicated displacement
Absolute expanded uncertainty : 0.7 µm
Dial gauge tester
Resolution : 0.001 mm
Approved on 22 March 2005
Institute service identifier : FTMC/8
Various dimensional , Hand Instruments : 0 mm to 2 mm
Dial gauge: error of indicated displacement
Absolute expanded uncertainty : 1.2 µm
Dial gauge tester
Resolution : 0.002 mm
Approved on 22 March 2005
Institute service identifier : FTMC/7
Various dimensional , Hand Instruments : 0 mm to 100 mm
Micrometer: error of indicated size
Absolute expanded uncertainty : 3.6 µm
Comparison to gauge block
Approved on 22 March 2005
Institute service identifier : FTMC/5
DC resistance , DC resistance standards and sources: low values : 0.1 Ω
Fixed resistor
Relative expanded uncertainty : 0.5 µΩ/Ω
Direct current comparator bridge
Power : 2.5 mW
Approved on 06 August 2013
Institute service identifier : FTMC/9
DC voltage (up to 1100 V) , DC voltage sources: low values : 0.1 V to 10 V
DC voltage source, multifunction calibrator
Relative expanded uncertainty : 2.5 µV/V to 1 µV/V
Direct measurement
Approved on 06 August 2013
Institute service identifier : FTMC/40
Frequency , Frequency : 1 Hz to 2.00E8 Hz
Rectangular pulse source
Relative expanded uncertainty : 6.7E-12 Hz/Hz
Direct measurement of frequency
Averaging time : 1000 s
Slewing rate : > 0.5 V/ns
Approved on 18 November 2009
Institute service identifier : FTMC/13

# Select Metrology Area

General physics
Acoustics, Ultrasound, Vibration

This metrology area covers the fields of acoustics, ultrasound and vibration.

Electricity and Magnetism

This metrology area covers the fields of DC and AC measurements, impedance, electric and magnetic fields, radiofrequencies and measurements on materials.

Length

This metrology area covers the fields of laser frequencies and dimensional metrology.

Mass and related quantities

This metrology area covers the fields of mass standards, force, pressure, density, hardness, torque, gravity and viscosity.

Chemistry and Biology
Chemistry and Biology

This metrology area covers all fields relevant to chemistry and biology.

Ionizing radiation
Ionizing Radiation

This metrology area covers the fields of dosimetry, radioactivity and neutron measurements.