Refine your search 

CMC AREA 



PHYSICS 




IONIZING RADIATION 




CHEMICAL MATERIAL 




CHEMICAL ANALYTE 




GEOGRAPHIC LOCATION 





Result of the search 

Your query 'china' produced more than 1000 results

New search

 Hong Kong, China, SCL (Standards and Calibration Laboratory)    Complete CMCs in Time and Frequency for Hong Kong, China (.PDF file)
  
Time scale difference. Local clock vs UTC, 1E09 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 1   Direct time interval measurement against UTC(SCL) using calibrated counter   1 PPS amplitude: 1 V to 5 V (50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/1 
 Time interval. Time difference source, 1E09 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 1   Direct time interval measurement using calibrated counter   Trigger level: 4 V to +4 V (50 ohm)   Slew rate: > 1E+09 V/s   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/21 
 Time interval. Time difference source, 1 s to 999 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 1 to 3   Direct time interval measurement using calibrated counter   Trigger level: 4 V to +4 V (50 ohm)   Slew rate: > 1E+09 V/s   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/22 
 Time interval. Time difference meter, 1E09 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 1   Measuring test signals from calibrated generator   Trigger level: 2 V to +2 V (50 ohm)   Slew rate: > 1E+09 V/s   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/28 
 Time interval. Time difference meter, 1 s to 999 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 1 to 15   Measuring test signals from calibrated generator   Trigger level: 2 V to +2 V (50 ohm)   Slew rate: > 1E+09 V/s   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/29 
 Frequency. Local frequency standard, 10 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 8.0E14   Phase comparison with laboratory frequency standard   Measurement time: 5 d to 10 d   Amplitude: 0.5 V to 3 V (rms into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/2 
 Frequency. Local frequency standard, 5 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 8.0E14   Phase comparison with laboratory frequency standard   Measurement time: 5 d to 10 d   Amplitude: 0.5 V to 3 V (rms into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/3 
 Frequency. Local frequency standard, 1 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 8.2E14   Phase comparison with laboratory frequency standard   Measurement time: 5 d to 10 d   Amplitude: 0.5 V to 3 V (rms into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/4 
 Frequency. Local frequency standard, 100 kHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 2.0E13   Phase comparison with laboratory frequency standard   Measurement time: 5 d to 10 d   Amplitude: 0.5 V to 3 V (rms into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/5 
 Frequency. General frequency source, 1 Hz to 1E+03 Hz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 4.0E06 to 3.0E09   Direct frequency measurement using calibrated frequency counter   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/6 
 Frequency. General frequency source, 1 kHz to 1E+03 kHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 3.0E09 to 5.0E11   Direct frequency measurement using calibrated frequency counter   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/7 
 Frequency. General frequency source, 1 MHz to 1.3E+03 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 5.0E11   Direct frequency measurement using calibrated frequency counter   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/8 
 Frequency. General frequency source, 1.3 GHz to 10 GHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 6.0E10 to 2.0E10   Direct frequency measurement using calibrated frequency counter   Gate time: 1 s   Number of measurements: 5 to 10   Amplitude: 0.01 mW to 3 mW (power into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/9 
 Frequency. General frequency source, 10 GHz to 26.5 GHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 2.0E10 to 1.0E10   Direct frequency measurement using calibrated frequency counter   Gate time: 1 s   Number of measurements: 5 to 10   Amplitude: 0.01 mW to 3 mW (power into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/10 
 Frequency. Frequency counter, 1 Hz to 1E+03 Hz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 4.0E06 to 3.0E09   Measuring test signals from calibrated frequency source   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/11 
 Frequency. Frequency counter, 1 kHz to 1E+03 kHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 3.0E09 to 6.1E11   Measuring test signals from calibrated frequency source   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/12 
 Frequency. Frequency counter, 1 MHz to 1.3E+03 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 6.1E11   Measuring test signals from calibrated frequency source   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/13 
 Frequency. Frequency counter, 1.3 GHz to 10 GHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 6.1E10 to 2.0E10   Measuring test signals from calibrated frequency source   Gate time: 1 s   Number of measurements: 5 to 10   Amplitude: 0.01 mW to 3 mW (power into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/14 
 Frequency. Frequency counter, 10 GHz to 26.5 GHz   Relative expanded uncertainty (k = 2, level of confidence 95 %): 2.0E10 to 1.0E10   Measuring test signals from calibrated frequency source   Gate time: 1 s   Number of measurements: 5 to 10   Amplitude: 0.01 mW to 3 mW (power into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/15 
 Time interval. Period source, 1 ms to 1E+03 ms   Relative expanded uncertainty (k = 2, level of confidence 95 %): 3.0E09 to 4.0E06   Derived from direct frequency measurement using calibrated counter   Gate time: 120 s to 1000 s   Number of measurements: 5 to 10   Amplitude: 0.3 V to 2.5 V (peaktopeak into 50 ohm)   Ideal DUT effect included Approved on 31 August 2005   Internal NMI service identifier: SCL/16 
