Refine your search 

CMC AREA 



PHYSICS 



IONIZING RADIATION 



GEOGRAPHIC LOCATION 





Result of the search 

Your query 'Norway' produced 134 results

New search

 Norway, JV (Justervesenet)    Complete CMCs in Photometry and Radiometry for Norway (.PDF file)
  
Responsivity, spectral, power. Broadband detector. Measurand unit, A/W, V/W, reading/W, W, dB, dBm   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 4.8   Double grating monochromator, cut in filters   Wavelength range: 260 nm to 400 nm   Bandwith: 4 nm   Power: 0.4 µW to 2.1 µW   Approved on 29 September 2001 
 Responsivity, spectral, power. Broadband detector. Measurand unit, A/W, V/W, reading/W, W, dB, dBm   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.63   Double grating monochromator, cut in filters   Wavelength range: 400 nm to 930 nm   Bandwith: 4 nm   Power: 0.3 µW to 1.5 µW   Approved on 29 September 2001 
 Responsivity, spectral, power. Broadband detector. Measurand unit, A/W, V/W, reading/W, W, dB, dBm   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 2.4   Double grating monochromator, cut in filters   Wavelength range: 930 nm to 1600 nm   Bandwith: 4 nm   Power: 1 µW to 6 µW   Approved on 29 September 2001 
 Mass. Mass standard, 1 mg to 100 mg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 0.6 to 1.5   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 0.1 g to 1 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 1.5 to 2.1   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 1 g to 10 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 2.1 to 9   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 10 g to 100 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 9 to 14   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 0.1 kg to 1 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 14 to 70   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 1 kg to 10 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 0.07 to 1.3   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 10 kg to 50 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 1.3 to 22   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 50 kg to 100 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 22 to 170   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass. Mass standard, 100 kg to 500 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in g: 0.17 to 1.1   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known. Approved on 02 December 2003 
 Mass flowrate. Flow meter, 0.1 kg/s to 25 kg/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.20   Coriolis, pulse or other electrical output   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO1 
 Mass flowrate. Flow meter, 0.1 kg/s to 25 kg/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.20   Coriolis, visual reading   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO2 
 Volume flowrate. Flow meter, 0.1 L/s to 25 L/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.22   Positive displacement, turbine, electromagnetic, ultrasonic, pulse or other electrical output   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO3 
 Volume flowrate. Flow meter, 0.1 L/s to 25 L/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.22   Positive displacement, turbine, electromagnetic, ultrasonic, visual reading   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO4 
 Mass. Mass meter, 1 kg to 5000 kg   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.10   Coriolis   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO5 
 Volume. Volume meter, 1 L to 5000 L   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.12   Positive displacement, turbine, electromagnetic, ultrasonic   Liquid: water   Temperature: 5 °C to 65 °C   Pipe size: DN 10  150   Approved on 02 December 2003   Internal NMI service identifier: NO6 
 Volume flowrate. Waterflow meter, 0.01 L/s to 4 L/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.20   Positive displacement, turbine, electromagnetic, ultrasonic   Liquid: water   Temperature: 5 °C to 15 °C   Pipe size: DN 5  50   Approved on 02 December 2003   Internal NMI service identifier: NO7 
 Volume. Water meter, 10 L to 500 L   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.10   Positive displacement, turbine, electromagnetic, ultrasonic   Liquid: water   Temperature: 5 °C to 15 °C   Pipe size: DN 5  50   Approved on 02 December 2003   Internal NMI service identifier: NO8 
