Refine your search 

CMC AREA 



PHYSICS 




IONIZING RADIATION 



CHEMICAL MATERIAL 




CHEMICAL ANALYTE 




GEOGRAPHIC LOCATION 





Result of the search 

Your query 'Netherlands' produced 756 results

New search

   Complete CMCs in Mass and related quantities for Netherlands (.PDF file)
  
Mass. Mass standard, 1 mg to 100 mg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 0.6 to 1.5   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 0.1 g to 1 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 1.5 to 3   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 1 g to 10 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 3 to 6   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 10 g to 100 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 6 to 15   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 0.1 kg to 1 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 15 to 100   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 1 kg to 10 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 0.1 to 1.5   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Mass. Mass standard, 10 kg to 50 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 1.5 to 50   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 03 January 2007 
 Absolute pressure. Pressure gauge, 5.0E+03 Pa to 3.5E+05 Pa   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in Pa: (0.2 + 3E05p), p pressure in Pa   Gas medium   Uncertainty values range from 3.5E01 Pa to 1.1E+01 Pa Approved on 03 January 2007 
 Gauge pressure. Pressure balance, 5.0E+03 Pa to 5.0E+05 Pa   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in Pa: (0.5 + 3E05p), p pressure in Pa   Gas medium   Uncertainty values range from 6.5E01 Pa to 1.6E+01 Pa Approved on 03 January 2007 
 Gauge pressure. Pressure balance, 5.0E+05 Pa to 2.0E+07 Pa   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in Pa: (1 + 4E05p), p pressure in Pa   Gas medium   Uncertainty values range from 2.1E+01 Pa to 8.0E+02 Pa Approved on 03 January 2007 
 Gauge pressure. Pressure balance, 1.0E+06 Pa to 8.0E+07 Pa   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in Pa: (5E05p), p pressure in Pa   Oil medium   Uncertainty values range from 5.0E+01 Pa to 4.0E+03 Pa Approved on 03 January 2007 
 Gauge pressure. Pressure balance, 8.0E+07 Pa to 5.0E+08 Pa   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in Pa: (1E04p), p pressure in Pa   Oil medium   Uncertainty values range from 8.0E+03 Pa to 5.0E+04 Pa Approved on 03 January 2007 
 Density of liquid. Liquids, 600 kg/m^{3} to 1000 kg/m^{3}   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.02   Several liquids   Approved on 03 January 2007   Internal NMI service identifier: VSL/NE24 
 Gas volume flow. Variable area, wetgas, LFE, massflow, soapfilmmeters, piston provers, 2.E05 m^{3}/h to 3.5 m^{3}/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.20 to 0.45   Direct or indirect calibration of volume per time   Gas: various gases   Pressure: < 1 MPa   Pipe size: various tubing   "Lowflow", conversion to kg/h possible on request Approved on 19 December 2012   Internal NMI service identifier: VSL/NE02 
 Gas volume flow. Turbine, rotary, thermal, mass Flow meters, 1 m^{3}/h to 400 m^{3}/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.09   Gas displacement per time   Gas: various gases   Pressure: 100 kPa   Pipe size: DN 10  DN 100   "Bell prover 3", conversion to kg/h possible on request Approved on 19 December 2012   Internal NMI service identifier: VSL/NE04 
 Gas volume flow. Turbine, rotary, US, vortexmeters and differential producers, 15 m^{3}/h to 15000 m^{3}/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.15   Master meter, pulse or analogue output of DuT   Gas: air   Pressure: 100 kPa   Pipe size: DN 80  DN 600   "Large testbench", conversion to kg/h possible on request Approved on 19 December 2012   Internal NMI service identifier: VSL/NE05 
 Gas volume flow. Rotary, turbine, ultrasonic, vortex, Coriolis meters and differential producers, 24500 m^{3}/h to 1.65E+06 m^{3}/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.3 to 0.22   Master meter, pulse or analogue output of DuT   Gas: natural gas   Pressure: 0.8 MPa to 6.3 MPa   Pipe size: 8" to 36" ANSI 600   "EuroLoop facility", Rotterdam Approved on 19 December 2012   Internal NMI service identifier: VSL/NE06 
 Gas volume flow. Rotary meters, 5 m^{3}/h to 230 m^{3}/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.29 to 0.06   Gas volume generation   Gas: several gases   Pressure: 0.1 MPa to 6.1 MPa   Pipe size: 4" or upon request   "Gas Oil Piston Prover" primary standard Approved on 19 December 2012   Internal NMI service identifier: VSL/NE08 
 Air velocity. Prantl and (s) Pitot tube, mechanical and thermal anemometer, 0.1 m/s to 1 m/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: [(3.2/v)  2.2], v velocity in m/s   Windtunnel   Gas: air   Pressure: 100 kPa   Tunnel diameter: 380 mm   Approved on 19 December 2012   Internal NMI service identifier: VSL/NE10 
 Air velocity. Prantl and (s) Pitot tube, mechanical and thermal anemometer, 1 m/s to 35 m/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 1   Windtunnel   Gas: air   Pressure: 100 kPa   Tunnel diameter: 380 mm   Approved on 19 December 2012   Internal NMI service identifier: VSL/NE10 
