Refine your search 

CMC AREA 



PHYSICS 



CHEMICAL MATERIAL 



CHEMICAL ANALYTE 



GEOGRAPHIC LOCATION 





Result of the search 

Your query 'Kazakhstan' produced 69 results

New search

 Kazakhstan, KazInMetr (Kazakh Institute of Metrology RSE)    Complete CMCs in Mass and related quantities for Kazakhstan (.PDF file)
  
Hardness. Hardness block, According to ISO 6508:1 and GOST 9013 HRA   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRA: 0.5   Rockwell A according ISO 6508:3 and GOST 9013   Preliminary test force: 98.07 N   Total test force: 588.4 N   Approved on 16 April 2010 
 Hardness. Hardness block, According to ISO 6501:3 and GOST 9013 HRB   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRB: 0.8   Rockwell B according ISO 6508:3 and GOST 9013   Preliminary test force: 98.07 N   Total test force: 980.7 N   Approved on 16 April 2010 
 Hardness. Hardness block, According to ISO 6508:1 and GOST 9013 HRC   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRC: 0.5   Rockwell C according ISO 6508:3 and GOST 9013   Preliminary test force: 98.07 N   Total test force: 1471 N   Approved on 16 April 2010 
 Hardness. Hardness block, According to ISO 6508:1 and GOST 22975 HRN   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRN: 0.7   Rockwell N according ISO 6508:3 and GOST 22975   Preliminary test force: 29.42 N   Total test force: 147.1 N to 441.3 N   Approved on 16 April 2010 
 Hardness. Hardness block, According to ISO 6508:3 and GOST 22975 HRT   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRT: 0.8   Rockwell 15T according ISO 6508:3 and GOST 22975   Preliminary test force: 29.42 N   Total test force: 147.1 N   Approved on 16 April 2010 
 Hardness. Hardness block, According to ISO 6508:3 and GOST 22975 HRT   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRT: 1.2   Rockwell 30T according ISO 6508:3 and GOST 22975   Preliminary test force: 29.42 N   Total test force: 294.2 N   Approved on 16 April 2010 
 Mass. Mass standards, 1 mg to 100 mg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 1.5 to 2.4   Subdivision method   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Mass. Mass standards, 0.1 g to 1 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 2.4 to 3   Subdivision method   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Mass. Mass standards, 1 g to 10 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 3 to 6   Subdivision method   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Mass. Mass standards, 10 g to 100 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 6 to 16   Subdivision method   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Mass. Mass standards, 0.1 kg to 1 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 16 to 120   Subdivision method   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Mass. Mass standards, 1 kg to 10 kg   Absolute expanded uncertainty (k = 2, level of confidence 95%) in mg: 0.12 to 1.6   Direct comparison   Temperature: 17.5 °C to 22.5 °C   Relative humidity: 40 % to 60 %   Uncertainty scales with measurand level. The volume of the mass standards is known Approved on 10 September 2014 
 Hardness. Hardness block, According to ISO 6507:1 HV   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 1.7   Vickers HV1 according to ISO 6507:3   Test force: 9.807 N   Approved on 28 February 2014 
 Hardness. Hardness block, According to ISO 6507:1 HV   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 1.5   Vickers HV5, HV30 according to ISO 6507:3   Test force: 49.03 N, 294.2 N   Approved on 28 February 2014 
 Viscosity. Newtonian liquids, 20 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.25   Reference liquids   Temperature: 20 °C 
 Viscosity. Newtonian liquids, 100 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.32   Reference liquids   Temperature: 20 °C 
 Viscosity. Newtonian liquids, 1000 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.4   Reference liquids   Temperature: 20 °C 
 Time scale difference. Local clock vs. UTC(KZ), 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95%) in ns: 5.1   Time interval measurement   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time scale difference. Local clock vs. UTC, 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95%) in ns: 151   Comparison against UTC   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Prediction time: 10 days   Predicted Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time scale difference. Remote clock vs. UTC(KZ), 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95%) in ns: 97   GPS common view comparison against UTC(KZ)   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Averaging time: 5 days   Distance: < 1500 km   Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
