Refine your search 

CMC AREA 



PHYSICS 



CHEMICAL MATERIAL 



CHEMICAL ANALYTE 



GEOGRAPHIC LOCATION 





Result of the search 

Your query 'Kazakhstan' produced 69 results

New search

 Kazakhstan, KazInMetr (Kazakh Institute of Metrology RSE)    Complete CMCs in Time and Frequency for Kazakhstan (.PDF file)
  
Time scale difference. Local clock vs UTC, 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 5.1   Time interval measurement   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time scale difference. Local clock vs UTC, 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 151   Comparison against UTC   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Prediction time: 10 days   Predicted Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time scale difference. Remote clock vs UTC, 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 97   GPS common view comparison against UTC(KZ)   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Averaging time: 5 days   Distance: < 1500 km   Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time scale difference. Remote clock vs UTC, 1 s to 1 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 172   GPS common view comparison against UTC   Pulse amplitude: > 1 V   Slew rate: > 0.5 V/ns   Prediction time: 10 days   Distance: < 1500 km   Predicted Fractional part of modulo 1 second difference, integer part can be provided also Approved on 13 March 2015 
 Time interval. Period source, 1E06 s to 1E+03 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 10   Time interval counter   Gate time: > 1 s   Slew rate: 0.5 V/ns   Approved on 13 March 2015 
 Time interval. Time difference source, 1E09 s to 1E06 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 2   Time interval measurement   Number of measurements: 100   Amplitude: 1 V to 4 V   Slew rate: 0.5 V/ns   Approved on 13 March 2015 
 Time interval. Time difference source, 1E06 s to 1E+03 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 2   Time interval measurement   Number of measurements: 10   Amplitude: 1 V to 4 V   Slew rate: 0.5 V/ns   Approved on 13 March 2015 
 Time interval. Time difference meter, 1E09 s to 1E06 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 2   Time interval measurement   Number of measurements: 100   Amplitude: 1 V to 4 V   Slew rate: 0.5 V/ns   Approved on 13 March 2015 
 Time interval. Time difference meter, 1E06 s to 1E+03 s   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in ns: 2   Time interval measurement   Number of measurements: 10   Amplitude: 1 V to 4 V   Slew rate: 0.5 V/ns   Approved on 13 March 2015 
 Frequency. Local frequency standard, 5 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 1E12   Frequency comparator   Averaging time: 1 s   Amplitude: 0.8 V to 1.2 V   Approved on 24 October 2011   Internal NMI service identifier: RP UKF "KazInMetr" 06 
 Frequency. Local frequency standard, 10 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 1E12   Frequency comparator   Averaging time: 1 s   Amplitude: 0.8 V to 1.2 V   Approved on 24 October 2011   Internal NMI service identifier: RP UKF "KazInMetr" 07 
 Frequency. Local frequency standard, 5 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 2.7E13   Frequency comparator   Averaging time: 1E+03 s   Amplitude: 0.8 V to 1.2 V   Approved on 24 October 2011   Internal NMI service identifier: RP UKF "KazInMetr" 08 
 Frequency. Local frequency standard, 10 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 2.7E13   Frequency comparator   Averaging time: 1E+03 s   Amplitude: 0.8 V to 1.2 V   Approved on 24 October 2011   Internal NMI service identifier: RP UKF "KazInMetr" 09 
 Frequency. Local frequency standard, 0.1 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 2.4E13   Phase measurement   Averaging time: 1 day   Amplitude: 0.5 V to 2.4 V   Approved on 13 March 2015 
 Frequency. Local frequency standard, 5 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 2.4E13   Phase measurement   Averaging time: 1 day   Amplitude: 0.5 V to 2.4 V   Approved on 13 March 2015 
 Frequency. Local frequency standard, 10 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 2.4E13   Phase measurement   Averaging time: 1 day   Amplitude: 0.8 V to 2.4 V   Approved on 13 March 2015 
 Frequency. Remote frequency standard, 5 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 1E12   GPS common view comparison against UTC   Averaging time: 5 days   Baseline length from KazInMetr: < 1500 km   Post processed Approved on 13 March 2015 
 Frequency. Remote frequency standard, 10 MHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 1E12   GPS common view comparison against UTC   Averaging time: 5 days   Baseline length from KazInMetr: < 1500 km   Post processed Approved on 13 March 2015 
 Frequency. General frequency source, 0.1 Hz to 1E+03 Hz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 3E07   Direct frequency measurement   Averaging time: 1000 s   Amplitude: > 0.5 Vrms   Approved on 13 March 2015 
 Frequency. General frequency source, 1 kHz to 1E+06 kHz   Relative expanded uncertainty (k = 2, level of confidence 95 %) in Hz/Hz: 3E09   Direct frequency measurement   Averaging time: 1000 s   Amplitude: > 0.5 Vrms   Approved on 13 March 2015 
