Refine your search 

CMC AREA 



PHYSICS 



IONIZING RADIATION 




CHEMICAL MATERIAL 




CHEMICAL ANALYTE 




GEOGRAPHIC LOCATION 





Result of the search 

Your query 'Greece' produced 177 results

New search

 Greece, EIM (Hellenic Institute of Metrology)    Complete CMCs in Mass and related quantities for Greece (.PDF file)
  
Mass. Mass standards, 1 mg to 100 mg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 0.6 to 1.5   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 0.1 g to 1 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 1.5 to 3   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 1 g to 10 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 3 to 5   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 10 g to 100 g   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 5 to 12   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 0.1 kg to 1 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in µg: 12 to 90   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 1 kg to 10 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 0.09 to 1.3   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 20 kg to 50 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 5.2 to 12   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 60 kg to 100 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in mg: 320 to 550   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Mass. Mass standards, 100 kg to 500 kg   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in g: 0.55 to 2.7   Comparison in air   Uncertainty scales with measurand level The volume of the mass standards is known Approved on 18 November 2010 
 Volume water flowrate. Flowmeters, 6 L/h to 30000 L/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.1   Gravimetric   Temperature: ambient to 70 °C   Pipe diameter: DN 10 to DN 50   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALFLO201 
 Volume gas flowrate. Flowmeters, 9 L/h to 72 L/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.2   Volumetric   Pipe diameter: DN 10   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALFLO100 
 Volume gas flowrate. Flowmeters, 72 L/h to 430 L/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.15   Volumetric   Pipe diameter: DN 10   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALFLO100 
 Volume gas flowrate. Flowmeters, 1000 L/h to 60000 L/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.2   Volumetric   Pipe diameter: DN 10 to DN 40   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALFLO100 
 Mass water flowrate. Flowmeters, 6 kg/h to 30000 kg/h   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.1   Gravimetric   Temperature: ambient to 70 °C   Pipe diameter: DN 10 to DN 50   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALFLO201 
 Volume of liquid. Micropipettes or piston pipettes, 20 µL to 1000 µL   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.2 to 0.05   Gravimetric   Liquid: water   Reference temperature: 20 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL101 
 Volume of liquid. Pipettes, burettes, 5 mL to 50 mL   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.02   Gravimetric   Liquid: water   Reference temperature: 20 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL101 
 Volume of liquid. Volumetric flasks, 10 mL to 5000 mL   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.1 to 0.015   Gravimetric   Liquid: water   Reference temperature: 20 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL101 
 Volume of liquid. Proving tanks, 1 L to 20 L   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.08 to 0.002   Gravimetric   Liquid: water   Reference temperature: 20 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL201 
 Volume of liquid. Proving tanks, 1 L to 20 L   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.01   Volumetric   Liquid: water   Reference temperature: 20 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL301 
 Volume of liquid. Proving tanks, 20 L to 400 L   Relative expanded uncertainty (k = 2, level of confidence 95 %) in %: 0.003 to 0.02   Gravimetric   Liquid: water   Reference temperature: 20 °C or 15 °C   Approved on 10 November 2011   Internal NMI service identifier: EIM/CALVOL202, EIM/CALVOL203 
