Refine your search 

CMC AREA 



PHYSICS 




IONIZING RADIATION 




CHEMICAL MATERIAL 




CHEMICAL ANALYTE 




GEOGRAPHIC LOCATION 





Result of the search 

Your query 'France' produced 912 results

New search

 France, LNE (Laboratoire national de métrologie et d'essais)    Complete CMCs in Acoustics, Ultrasound, Vibration for France (.PDF file)
  
Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.03   IEC 610942:1992   Frequency: 50 Hz to 2 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.035   IEC 610942:1992   Frequency: 2.5 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.045   IEC 610942:1992   Frequency: 3.15 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.055   IEC 610942:1992   Frequency: 4 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.075   IEC 610942:1992   Frequency: 5 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.10   IEC 610942:1992   Frequency: 6.3 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.14   IEC 610942:1992   Frequency: 8 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS1P. Measurand unit, dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.20   IEC 610942:1992   Frequency: 10 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.035   IEC 610942:1992   Frequency: 50 Hz to 3.15 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.04   IEC 610942:1992   Frequency: 4 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.045   IEC 610942:1992   Frequency: 5 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.05   IEC 610942:1992   Frequency: 6.3 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.065   IEC 610942:1992   Frequency: 8 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.085   IEC 610942:1992   Frequency: 10 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.11   IEC 610942:1992   Frequency: 12.5 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.15   IEC 610942:1992   Frequency: 16 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type LS2aP. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.20   IEC 610942:1992   Frequency: 20 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type WS1. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.06   Substitution in closed coupler   Frequency: 50 Hz to 2 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type WS1. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.09   Substitution in closed coupler   Frequency: 2.5 kHz to 5 kHz   Approved on 18 October 2005 
 Pressure sensitivity level. Measurement microphone type WS1. Measurand unit, to dB (reference: 1 V/Pa)   Absolute expanded uncertainty (k = 2, level of confidence 95%) in dB: 0.15   Substitution in closed coupler   Frequency: 6.3 kHz to 10 kHz   Approved on 18 October 2005 
