Refine your search 

CMC AREA 



PHYSICS 




IONIZING RADIATION 




CHEMICAL MATERIAL 




CHEMICAL ANALYTE 




GEOGRAPHIC LOCATION 





Result of the search 

Your query 'United Kingdom' produced more than 1000 results

New search

 United Kingdom, NPL (National Physical Laboratory)    Complete CMCs in Photometry and Radiometry for United Kingdom (.PDF file)
  
Colour, surface, L*a*b*. Fluorescent material, a*, b* = 200 to a*, b* = 200   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 1.2   Reference spectrofluorimeter   Irradiation conditions: monochromator   Specific measurement conditions: 0/45   Type of fluorescent material: general   Approved on 07 May 2007 
 Colour, surface, L*a*b*. Diffusely reflecting material, L* = 0 to L* = 100   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.15   Integrating sphere spectrophotometer   Specific measurement conditions: 0/d, 2 nm bandwidth   Other types of material can also be calibrated Approved on 07 May 2007 
 Colour, surface, L*a*b*. Diffusely reflecting material, a*, b* = 200 to a*, b* = 200   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.1   Integrating sphere spectrophotometer   Specific measurement conditions: 0/d, 2 nm bandwidth   Other types of material can also be calibrated Approved on 07 May 2007 
 Colour, transmitted, x, y, Y. General material, x, y = 0 to x, y = 0.9   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.0002   Spectrophotometer   Specific measurement conditions: regular transmittance geometry   Type of material: nonfluorescent 
 Colour, transmitted, x, y, Y. General material, Y = 0 to Y = 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.003   Spectrophotometer   Specific measurement conditions: regular transmittance geometry   Type of material: nonfluorescent 
 Colour, transmitted, L*a*b*. General material, L* = 0 to L* = 100   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.05   Spectrophotometer   Specific measurement conditions: regular transmittance geometry   Type of material: nonfluorescent 
 Colour, transmitted, L*a*b*. General material, a*, b* = 200 to a*, b* = 200   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.05   Spectrophotometer   Specific measurement conditions: regular transmittance geometry   Type of material: nonfluorescent 
 Wavelength. Fibre optic source, 800 nm to 1700 nm   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in nm: 0.3   Spectrum analyser 
 Wavelength. Optical spectrum analyser, 600 nm to 1700 nm   Absolute expanded uncertainty (k = 2, level of confidence 95 %) in pm: 34   Lamp / wavemeter / transition locked laser   Measurements can be made at specified wavelengths within this range, depending on the availability of a suitable laser source. The uncertainty will depend on the properties of the OSA under calibration. 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 3E05 to 3.5E04, varies as the logarithm of the measurand   Reference spectrophotometer   Wavelength: 250 nm to 749 nm   Bandwidth: < 3 nm 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 1E05 to 4.5E04, varies as the logarithm of the measurand   Reference spectrophotometer   Wavelength: 750 nm to 2500 nm   Bandwidth: < 3 nm 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.0017 at 0.73   Scanning spectrophotometer   Wavenumber: 3990 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.00065 at 0.14   Scanning spectrophotometer   Wavenumber: 3512 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.0011 at 0.36   Scanning spectrophotometer   Wavenumber: 3031 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.00052 at 0.06   Scanning spectrophotometer   Wavenumber: 2739 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.00079 at 0.17   Scanning spectrophotometer   Wavenumber: 2598 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.00064 at 0.10   Scanning spectrophotometer   Wavenumber: 2473 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, regular, spectral. Spectrallyneutral material, 1E03 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.0001 at 0.0001   Scanning spectrophotometer   Wavenumber: 2010 cm^{1}   Bandwidth: 2 cm^{1} to 15 cm^{1}   Specific measurement conditions: measured at 5 ° and f/6. Values calculated for normal incidence and a range of f/n's.   Other types of material can also be calibrated. Uncertainties at other transmittance values available on request. Approved on 07 May 2007 
 Transmittance, diffuse, spectral. Spectrallyneutral material, 0 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.01   Spectrophotometer with integrating sphere accessory   Wavelength range: 200 nm to 800 nm   Bandwidth: 0.5 nm to 2 nm   Specific measurement conditions: 0/d   Other types of material can also be calibrated 
 Transmittance, diffuse, spectral. Spectrallyneutral material, 0 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95 %): 0.01   Spectrophotometer with integrating sphere accessory   Wavelength range: 800 nm to 2500 nm   Bandwidth: 2 nm to 20 nm   Specific measurement conditions: 0/d   Other types of material can also be calibrated 
