Refine your search 

CMC AREA 



PHYSICS 



GEOGRAPHIC LOCATION 





Result of the search 

 Japan, NMIJ (National Metrology Institute of Japan)    Complete CMCs in Mass and related quantities for Japan (.PDF file)
  
Hardness. Hardness block, 20 HRC to 60 HRC   Absolute expanded uncertainty (k = 2, level of confidence 95%) in HRC: 0.3   Rockwell C according to ISO 65083   Total test force: 1471 N   Approved on 24 March 2007   Internal NMI service identifier: NMIJ/MHR01 
 Torque. Torque measuring device, 0.005 kN m to 1 kN m   Relative expanded uncertainty (k = 2, level of confidence 95%): 5.0E05   Direct comparison   Mode: clockwise, anticlockwise   Approved on 24 March 2007   Internal NMI service identifier: NMIJ/MTQ11 
 Torque. Torque measuring device, 1 kN m to 20 kN m   Relative expanded uncertainty (k = 2, level of confidence 95%): 7.0E05   Direct comparison   Mode: clockwise, anticlockwise   Approved on 24 March 2007   Internal NMI service identifier: NMIJ/MTQ21 
 Torque. Reference torque wrench, 0.005 kN m to 1 kN m   Relative expanded uncertainty (k = 2, level of confidence 95%): 7.0E05   Direct comparison   Mode: clockwise, anticlockwise   Approved on 24 March 2007   Internal NMI service identifier: NMIJ/MTQ12 
 Kinematic viscosity. Newtonian liquids, 0.5 mm^{2}/s to 1.8 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.04   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk11 
 Kinematic viscosity. Newtonian liquids, 1.8 mm^{2}/s to 5.0 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.05   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk12 
 Kinematic viscosity. Newtonian liquids, 5.0 mm^{2}/s to 20 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.06   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk13 
 Kinematic viscosity. Newtonian liquids, 20 mm^{2}/s to 150 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.07   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk14 
 Kinematic viscosity. Newtonian liquids, 150 mm^{2}/s to 940 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.09   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk15 
 Kinematic viscosity. Newtonian liquids, 940 mm^{2}/s to 2000 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.10   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk16 
 Kinematic viscosity. Newtonian liquids, 2000 mm^{2}/s to 14000 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.12   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk17 
 Kinematic viscosity. Newtonian liquids, 14000 mm^{2}/s to 52000 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.14   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk18 
 Kinematic viscosity. Newtonian liquids, 52000 mm^{2}/s to 500000 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.17   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk19 
 Kinematic viscosity. Newtonian liquids, 0.5 mm^{2}/s to 1.2 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.05   Viscosity measurement   Temperature: 100 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk21 
 Kinematic viscosity. Newtonian liquids, 1.2 mm^{2}/s to 4.0 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.05   Viscosity measurement   Temperature: 100 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk22 
 Kinematic viscosity. Newtonian liquids, 4.0 mm^{2}/s to 20 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.06   Viscosity measurement   Temperature: 100 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk23 
 Kinematic viscosity. Newtonian liquids, 20 mm^{2}/s to 40 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.07   Viscosity measurement   Temperature: 100 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk24 
 Kinematic viscosity. Newtonian liquids, 0.5 mm^{2}/s to 80 mm^{2}/s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.07   Viscosity measurement   Temperature:  40 ºC   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVk31 
 Dynamic viscosity. Newtonian liquids, 0.4 mPa s to 1.4 mPa s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.07   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVd11 
 Dynamic viscosity. Newtonian liquids, 1.4 mPa s to 4.1 mPa s   Relative expanded uncertainty (k = 2, level of confidence 95%) in %: 0.07   Viscosity measurement   Temperature: 40 °C to 20 °C   The uncertainty of the viscosity of water (ISO/TR 3666 (1998), 0.17 %) is not taken into account Approved on 31 March 2011   Internal NMI service identifier: NMIJ/FFLVd12 
