Home Key and supplementary comparisons Calibration and Measurement Capabilities - CMCs
 KCDB home  > Free search results
 The BIPM key comparison database
CMC AREA
 CMCs General Physics (27)
PHYSICS
DC voltage, current, and resistance (27)
 DC resistance, DC resistance standards and sources, intermediate values (6)
 DC current, DC current sources, intermediate values (5)
 DC current, DC current meters, intermediate values (5)
 DC resistance, DC resistance meters, intermediate values (4)
 DC voltage, DC voltage sources, intermediate values (2)
 DC resistance, DC resistance meters, high values (2)
 DC resistance, DC resistance standards and sources, low values (1)
 DC resistance, DC resistance meters, low values (1)
 DC current, DC current sources, low values (1)
DC current, DC current meters, low values
DC voltage, DC voltage meters, intermediate values
DC voltage, DC voltage sources, low value ranges
DC resistance, DC resistance standards and sources, high values
GEOGRAPHIC LOCATION
SIM (27)
 Costa Rica (27)
 Result of the search
 Your query 'Costa Rica' produced 27 results New search
 1 2

 Costa Rica, ICE-LMVE (Instituto Costarricense de Electricidad, Laboratorio Metrológico de Variables Eléctricas)
Complete CMCs in Electricity and Magnetism for Costa Rica (.PDF file)
DC resistance, DC resistance meters, intermediate values. Multimeter, 1.9 to 100000000
Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 8.7 to 102
Direct comparison
Resistance, fixed values: (1.9, 10, 19, 100, 190) , (1, 1.9, 10, 19, 100, 190) k, (1, 1.9, 10, 19, 100) M
Approved on 23 March 2010
Internal NMI service identifier: ICE-LMVE/22201
Uncertainty table
 DC voltage, DC voltage sources, intermediate values. DC voltage source, multifunction calibrator, > 10 V to 100 V Relative expanded uncertainty (k = 2, level of confidence 95%) in µV/V: 4.9 to 6.4 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/11301
 DC voltage, DC voltage sources, intermediate values. DC voltage source, multifunction calibrator, 100 V to 1000 V Relative expanded uncertainty (k = 2, level of confidence 95%) in µV/V: 6.4 to 15.6 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/11302
 DC resistance, DC resistance standards and sources, low values. Fixed resistor, resistance box, 0.1 to 1 Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 53 to 402 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21101
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, > 1 to 10 Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 18 to 53 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21201
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, 10 to 100 Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 16 to 50 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21202
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, 100 to 1000 Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 10 to 14 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21203
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, 1 k to 10 k Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 10 to 14 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21204
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, 10 k to 100 k Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 10 to 14 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21205
 DC resistance, DC resistance standards and sources, intermediate values. Fixed resistor, resistance box, 100 k to 1000 k Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 16 to 30 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/21206
 DC resistance, DC resistance meters, low values. Multimeter, 1 Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 97 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22101
 DC resistance, DC resistance meters, intermediate values. Teraohmmeter, 10 M Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 286 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22202
 DC resistance, DC resistance meters, intermediate values. Teraohmmeter, 100 M Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 174 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22203
 DC resistance, DC resistance meters, intermediate values. Teraohmmeter, 1 G Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 235 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22204
 DC resistance, DC resistance meters, high values. Teraohmmeter, 10 G Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 694 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22301
 DC resistance, DC resistance meters, high values. Teraohmmeter, 100 G Relative expanded uncertainty (k = 2, level of confidence 95%) in µ/: 929 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/22302
 DC current, DC current sources, low values. Multifunction calibrator, current generator, 0.01 mA to 0.1 mA Relative expanded uncertainty (k = 2, level of confidence 95%) in µA/A: 26 to 82 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/31101
 DC current, DC current sources, intermediate values. Multifunction calibrator, current generator, > 0.1 mA to 1 mA Relative expanded uncertainty (k = 2, level of confidence 95%) in µA/A: 23 to 59 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/31201
 DC current, DC current sources, intermediate values. Multifunction calibrator, current generator, 1 mA to 10 mA Relative expanded uncertainty (k = 2, level of confidence 95%) in µA/A: 23 to 59 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/31202
 DC current, DC current sources, intermediate values. Multifunction calibrator, current generator, 10 mA to 100 mA Relative expanded uncertainty (k = 2, level of confidence 95%) in µA/A: 35 to 70 Direct comparison Approved on 23 March 2010 Internal NMI service identifier: ICE-LMVE/31203